Tang, T. L.*; Uesaka, Tomohiro*; Kawase, Shoichiro; Beaumel, D.*; Dozono, Masanori*; Fujii, Toshihiko*; Fukuda, Naoki*; Fukunaga, Taku*; Galindo-Uribarri. A.*; Hwang, S. H.*; et al.
Physical Review Letters, 124(21), p.212502_1 - 212502_6, 2020/05
The structure of a neutron-rich F nucleus is investigated by a quasifree () knockout reaction. The sum of spectroscopic factors of orbital is found to be 1.0 0.3. The result shows that the O core of F nucleus significantly differs from a free O nucleus, and the core consists of 35% O, and 65% excited O. The result shows that the O core of F nucleus significantly differs from a free O nucleus. The result may infer that the addition of the proton considerably changes the neutron structure in F from that in O, which could be a possible mechanism responsible for the oxygen dripline anomaly.
Oguri, Hidetomo; Okoshi, Kiyonori; Ikegami, Kiyoshi*; Takagi, Akira*; Asano, Hiroyuki; Shibata, Takanori*; Nammo, Kesao*; Ueno, Akira; Shinto, Katsuhiro
AIP Conference Proceedings 1869, p.030053_1 - 030053_7, 2017/08
A cesiated RF-driven negative hydrogen ion source was started to operate in September, 2014 in response to the need for upgrading J-PARC's linac beam current. The ion source mainly comprises a stainless-steel plasma chamber, a beam extractor, and a large vacuum chamber with two turbo molecular pumps of 1500 L/s for differential pumping. The ion source has been successfully providing the required beam current to the accelerator without any significant issues other than a single-incident antenna failure occurred in October, 2014. Continuous operation for approximately 1,000 h was achieved with a beam current and duty factor of 45 mA and 1.25 % (0.5 msec and 25 Hz), respectively. In this paper, we will present the some operation parameters and the beam stability through the long-term user operation.
Shibata, Takanori*; Asano, Hiroyuki; Ikegami, Kiyoshi*; Naito, Fujio*; Nammo, Kesao*; Oguri, Hidetomo; Okoshi, Kiyonori; Shinto, Katsuhiro; Takagi, Akira*; Ueno, Akira
AIP Conference Proceedings 1869, p.030017_1 - 030017_11, 2017/08
From September 2014, operation of Cs-seeded, multi-cusp, Radio Frequency (RF), hydrogen negative ion source (J-PARC source) has been started. The operation for 1,000 hours of J-PARC source has been achieved with H beam current 45 mA and duty factor of 1.25 % (0.5 msec and 25 Hz). In the present study, mechanisms of hydrogen plasma ramp-up and H production/transport processes in the steady state (which lasts for few 100 us) are investigated by numerical modeling for RF plasma. In the simulation, charged particle (e, H, H, and Cs) transport, time variations of inductive and capacitive electromagnetic field, collision processes between charged and neutral (H, H) particles are solved simultaneously. The model is applied to KEK parallel computation System-A with 32 nodes and 256 GB memory in order to solve high density RF plasma up to around 10 m with adequate statisticity. In the presentation, time variations of plasma density distributions and average energy are shown with electromagnetic field variations.
Shibata, Takanori*; Ikegami, Kiyoshi*; Liu, Y.*; Maruta, Tomofumi*; Naito, Fujio*; Takagi, Akira*; Asano, Hiroyuki; Kondo, Yasuhiro; Miura, Akihiko; Oguri, Hidetomo; et al.
Proceedings of 28th International Linear Accelerator Conference (LINAC 2016) (Internet), p.251 - 253, 2017/05
After upgrade of J-PARC Linac in 2014, Low Energy Beam Transport (LEBT) beam commissioning of the J-PARC LINAC has been made for improving H beam intensity extracted from Linac. Currents of two solenoid coils and steering magnets in LEBT are optimized with extraction and acceleration voltages for static acceleration in ion source (IS) which decides on an initial emittance diagram of H beam. As a result of LEBT and IS parameter optimization, beam transmission rate of RFQ has been reached up to 96% in 50 mA H current operation. Moreover, PIC-MC (Particle-In-Cell Monte-Carlo) simulation models developed for H transport in LEBT. Comparison between experimental and numerical results are presented to clarify beam physics from IS exit to RFQ entrance.
Okoshi, Kiyonori; Ikegami, Kiyoshi*; Takagi, Akira*; Asano, Hiroyuki; Ueno, Akira; Shibata, Takanori*; Nammo, Kesao*; Shinto, Katsuhiro; Oguri, Hidetomo
Proceedings of 13th Annual Meeting of Particle Accelerator Society of Japan (Internet), p.940 - 943, 2016/11
In 2014 October, operation of cesium-seeded Radio Frequency (RF)-driven negative hydrogen ion (H) source was started in J-PARC LINAC. Due to the skillful RF antenna screening and the proper pre-conditioning process, long term unscheduled beam stop due to the antenna failure did not occur for more than 1 year. The continuous operation of 1,350 hours with a peak beam current of 45 mA was achieved in the recent beam run. Moreover, the fluctuation of beam current in a low energy beam transport (LEBT) is kept within 2% of target value by application of three feedback systems. An ion source test-stand was assembled to investigate the ion source beam characteristic, for example, the emittance at the position of the RFQ entrance. In the presentation, we will report the recent status of the ion source operation and some experimental results obtained at the ion source test-stand.
Hirano, Koichiro; Asano, Hiroyuki; Ishiyama, Tatsuya; Ito, Takashi; Okoshi, Kiyonori; Oguri, Hidetomo; Kondo, Yasuhiro; Kawane, Yusuke; Kikuzawa, Nobuhiro; Sato, Yoshikatsu; et al.
Proceedings of 13th Annual Meeting of Particle Accelerator Society of Japan (Internet), p.310 - 313, 2016/11
We have used a beam scraper with the incident angle of 65deg to reduce the beam power deposition density in the MEBT between a 324 MHz RFQ and a 50-MeV DTL of the J-PARC linac. The 65 scraper was irradiated by the H beam up to particle number of 1.47E22. We observed a lot of surface projections with several hundred micrometers high in the beam irradiation damage on the scraper by using the laser microscope. In order to study the limits of scrapers, we constructed a new 3 MeV linac at J-PARC. We will conduct the scraper irradiation test at the end of this year.
Ueno, Akira; Okoshi, Kiyonori; Ikegami, Kiyoshi*; Takagi, Akira*; Asano, Hiroyuki; Oguri, Hidetomo
Review of Scientific Instruments, 87(2), p.02B130_1 - 02B130_5, 2016/02
The Japan Proton Accelerator Research Complex (J-PARC) cesiated rf-driven H ion source (IS), whose requirements are a peak beam intensity of 60mA within normalized emittances of 1.5 mmmrad both horizontally and vertically, a flat top beam duty factor of 1.25% (500 s 25 Hz) and a life-time of longer than 1month, has been successfully operated for about one year. The results of the fine-tuning to minimize the emittances of the J-PARC-IS with plasma chamber #3, which had the largest emittances with initial settings among four plasma chambers, will be presented in this paper. The rod-filter-filed will be finely tuned by selecting magnets with slightly different field strengths and/or changing gap-lengths. The dependence of the beam-hole-diameter on the emittances will be also presented. The tuning procedure to improve the emittances is one of the most important technologies for the IS of the high-energy and high-intensity accelerator.
Oguri, Hidetomo; Okoshi, Kiyonori; Ikegami, Kiyoshi*; Takagi, Akira*; Asano, Hiroyuki; Ueno, Akira; Shibata, Takanori*
Review of Scientific Instruments, 87(2), p.02B138_1 - 02B138_3, 2016/02
For the upgrade of the Japan Proton Accelerator Research Complex (J-PARC) linac beam current, a cesiated RF-driven negative hydrogen ion source was installed in 2014 summer shutdown period, and started to operate on September 29, 2014. The ion source has been successfully operated with a beam current and a duty factor of 33 mA and 1.25% (0.5 ms and 25 Hz), respectively. The result of recent beam operation showed that the ion source is capable of continuous operation for approximately 1,100 h. The spark rate at the beam extractor was observed to be less than once a day, which is acceptable level for the user operation. Although the antenna failure occurred during the user operation on October 26, 2014, there were no further serious troubles since then. In this conference, we will present the some operation parameters and the beam stability of the RF-driven ion source through the long-term user operation.
Shibata, Takanori*; Nishida, Kenjiro*; Mochizuki, Shintaro*; Mattei, S.*; Lettry, J.*; Hatayama, Akiyoshi*; Ueno, Akira; Oguri, Hidetomo; Okoshi, Kiyonori; Ikegami, Kiyoshi*; et al.
Review of Scientific Instruments, 87(2), p.02B128_1 - 02B128_3, 2016/02
A numerical model of plasma transport and electromagnetic field in the J-PARC RF ion source has been developed to understand relation between antenna coil heat loadings and plasma production/transport processes. From the calculation, the local plasma density increase is observed in the region close to the antenna coil. The magnetic field line with absolute magnetic flux density 30-120 Gauss results in the magnetization of electron which leads to high local ionization rate. The results suggest that modification of magnetic configuration can be made to reduce plasma heat flux onto the antenna.
Okoshi, Kiyonori; Ueno, Akira; Ikegami, Kiyoshi*; Takagi, Akira*; Asano, Hiroyuki; Oguri, Hidetomo
Proceedings of 12th Annual Meeting of Particle Accelerator Society of Japan (Internet), p.1040 - 1044, 2015/09
In order to satisfy the requirements of the J-PARC second stage, which will produce a 1MW beam, the Cs-free LaB-filament-driven H ion source was replaced to a cesiated RF-driven H ion source (RF ion source) during the summer shutdown period in 2014. By developing and testing the RF ion source on an off-line test-stand for about five years, it successfully and stably produced a peak current of more than 60mA within the acceptance of the following RFQ linac with the required beam duty-factor of more than 1.25% (500s 25Hz). For about one year, the RF ion source has been operated with a peak current of 33mA routinely, which causes low beam loss in the J-PARC operated with a 500kW beam. It has been also operated with a peak current of 55mA for a few weeks in each two months for the J-PARC beam study striving toward the 1MW beam operation. The beam current is controlled within the error of 0.1mA by feed-backing the RF power and the Cs density during the operation. Except for the only one failure of the internal RF-antenna in the beginning of the operation, there was no serious trouble. The RF ion source with the 33mA peak current operation is successfully operated for approximately 1100 hours.
Tamura, Jun; Ao, Hiroyuki; Nemoto, Yasuo; Asano, Hiroyuki*; Suzuki, Takahiro*
Journal of the Korean Physical Society, 66(3), p.399 - 404, 2015/02
ACSs are going to increase the beam energy of the J-PARC linac from 181 to 400 MeV. The 2011 Tohoku earthquake damaged the J-PARC facilities and the cavity conditioning was interrupted for two years. After the restoration of the facility, the two ACS cavities (M01 and M11) were conditioned. They achieved to 15-20% above the designed accelerating field of 4.2 MV/m. Since M01 was conditioned six years ago, the conditioning time required for M01 was drastically reduced from that for the first time. During the high-power operation for M11, which is an unique cavity equipped with a capacitive iris in a waveguide, no significant increase of the temperature and the discharge rate around the capacitive iris was observed. The vacuum pressure was sufficiently low (410 Pa) even from the view point of beam loss due to residual gas scattering. The more stable operation can be expected through one-month conditioning before beam commissioning.
Tamura, Jun; Ao, Hiroyuki; Nemoto, Yasuo; Asano, Hiroyuki*; Suzuki, Takahiro*
Proceedings of 10th Annual Meeting of Particle Accelerator Society of Japan (Internet), p.437 - 439, 2014/06
no abstracts in English
Katsuyama, Chie*; Nashimoto, Hiroaki*; Nagaosa, Kazuyo*; Ishibashi, Tomotaka*; Furuta, Kazuki*; Kinoshita, Takeshi*; Yoshikawa, Hideki; Aoki, Kazuhiro; Asano, Takahiro*; Sasaki, Yoshito; et al.
FEMS Microbiology Ecology, 86(3), p.532 - 543, 2013/12
Anaerobic microbial activity has a major influence on the subsurface environment, and should be considered in subsurface activities including the construction of radioactive waste repositories. We investigated denitrification and methanogenesis in anoxic groundwater from 140 m depth in two boreholes, where the redox potential fluctuated. The average maximum potential denitrification rates, measured under anaerobic conditions in the two boreholes using an N tracer. Methanogenesis candidates were detected by 16S rRNA gene analysis. Although the stable isotope signatures suggested that some of the dissolved methane was of biogenic origin, no potential for methane production was evident during the incubations. The groundwater at 140 m depth did not contain oxygen, had an Eh ranging from -144 to 6.8 mV, and was found to be a potential field for denitrification.
Ao, Hiroyuki; Asano, Hiroyuki; Tamura, Jun; Ouchi, Nobuo; Takata, Koji*; Naito, Fujio*
Proceedings of 9th Annual Meeting of Particle Accelerator Society of Japan (Internet), p.1124 - 1126, 2013/08
In the case of the RF window operated with the standing-wave cavity coupled to waveguides using coupling apertures, the reflection of the RF window affects the total coupling factor (voltage standing wave ratio, VSWR) consisted of the RF window and the cavity. Hence, the reflection (VSWR) of the RF window should be minimized not to change the cavity VSWR optimized separately, and to prevent standing waves between the RF window and the cavity. For minimizing the VSWR of the pillbox-type RF window, this report proposes that the dielectric constant of the ceramic is measured directly using the resonant frequency in the fabrication process of the RF window. In this work, three RF windows are fabricated additionally using the same type ceramic (NTK HA95), the small VSWR could be achieved using this procedure. This result also shows that the dielectric constant increases linearly with increasing the density.
Tamura, Jun; Ao, Hiroyuki; Asano, Hiroyuki; Ikegami, Masanori*; Maruta, Tomofumi; Miura, Akihiko; Morishita, Takatoshi; Oguri, Hidetomo; Ouchi, Nobuo; Sawabe, Yuki*; et al.
Proceedings of 9th Annual Meeting of Particle Accelerator Society of Japan (Internet), p.377 - 380, 2013/08
In the J-PARC linac, almost all the cavities and magnets have been precisely aligned because the accelerator tunnel has been deformed by the 2011 Tohoku Earthquake. In the beam transport line called MEBT2 and A0BT at the downstream of the drift tube linac, the beam duct has been roughly aligned after the precise alignment of the quadruple doublets. During the first beam operation after the earthquake, remarkable beam loss and residual radiation have been recognized at the MEBT2 and A0BT. As the result of the duct position measurement, the misalignement including over 10 mm shift from the beam axis was found. By conducting the beam duct realignment, the beam loss and the residual radiation were successfully decreased. In this paper, the procedure of the beam duct alignment and the beam loss due to the misalignment are described.
Takano, Katsuyoshi*; Asano, Atsushi*; Maeyoshi, Yuta*; Marui, Hiromi*; Omichi, Masaaki*; Saeki, Akinori*; Seki, Shu*; Sato, Takahiro; Ishii, Yasuyuki; Kamiya, Tomihiro; et al.
Journal of Photopolymer Science and Technology, 25(1), p.43 - 46, 2012/07
Takano, Katsuyoshi*; Sugimoto, Masaki; Asano, Atsushi*; Maeyoshi, Yuta*; Marui, Hiromi*; Omichi, Masaaki*; Saeki, Akinori*; Seki, Shu*; Sato, Takahiro; Ishii, Yasuyuki; et al.
Transactions of the Materials Research Society of Japan, 37(2), p.237 - 240, 2012/06
Maeyoshi, Yuta*; Takano, Katsuyoshi*; Asano, Atsushi*; Marui, Hiromi*; Omichi, Masaaki*; Sato, Takahiro; Kamiya, Tomihiro; Ishii, Yasuyuki; Okubo, Takeru; Koka, Masashi; et al.
Japanese Journal of Applied Physics, 51(4R), p.045201_1 - 045201_4, 2012/04
Morishita, Takatoshi; Asano, Hiroyuki; Ikegami, Masanori*
Proceedings of 2nd International Particle Accelerator Conference (IPAC '11) (Internet), p.2601 - 2603, 2011/09
The alignment has seriously been damaged due to the large earthquake on March 11th, 2011 in eastern Japan. There has been observed more than 40 mm settlement and about 25 mm horizontal deformation at the straight section in the linac accelerator tunnel. After that earthquake, there was continuous settlement in the straight section for two months. The linac accelerator tunnel has two expansion joints at the transport section to RCS. At these joints, abrupt displacements have been observed more than several millimeters both horizontally and vertically. Now, realignment started for restoration of the linac alignment.
Ao, Hiroyuki; Asano, Hiroyuki; Tamura, Jun; Ouchi, Nobuo; Hasegawa, Kazuo; Morishita, Takatoshi; Takata, Koji*; Naito, Fujio*; Yamazaki, Yoshishige
Proceedings of 8th Annual Meeting of Particle Accelerator Society of Japan (Internet), 3 Pages, 2011/08
An annular-ring coupled structure (ACS) for J-PARC linac has been in mass-production since March 2009. It was planned to fabricate 18 ACS accelerating modules and two debuncher modules for the three years. At present, they are progressing on schedule. The past achievements include: ten accelerating modules were completed on schedule; the frequency tuning of the most accelerating modules was finished except two debencher modules; and 14 accelerating modules have been brazed. One of the mass-produced modules was conditioned up to 1.6 MW, which corresponds to the accelerating field by 15% higher than the designed value of 4.2 MV/m.