Otsuka, Satoshi; Tanno, Takashi; Oka, Hiroshi; Yano, Yasuhide; Tachi, Yoshiaki; Kaito, Takeji; Hashidate, Ryuta; Kato, Shoichi; Furukawa, Tomohiro; Ito, Chikara; et al.
2018 GIF Symposium Proceedings (Internet), p.305 - 314, 2020/05
Oxide Dispersion Strengthened (ODS) steel has been developed worldwide as a high-strength and radiation-tolerant steel used for advanced nuclear system. Japan Atomic Energy Agency (JAEA) has been developing ODS steel as the primary candidate material of Sodium cooled Fast Reactor (SFR) high burn-up fuel cladding tube. Application of high burn-up fuel to SFR core can contribute to improvement of economical performance of SFR in conjunction with volume and hazardousness reduction of radioactive waste. This paper described the current status and future prospects of ODS tempered martensitic steel development in JAEA for SFR fuel application.
Ito, Chikara; Naito, Hiroyuki; Ishikawa, Takashi; Ito, Keisuke; Wakaida, Ikuo
JPS Conference Proceedings (Internet), 24, p.011038_1 - 011038_6, 2019/01
A high-radiation resistant optical fiber has been developed in order to investigate the interiors of the reactor pressure vessels and the primary containment vessels at the Fukushima Daiichi Nuclear Power Station. The tentative dose rate in the reactor pressure vessels is assumed to be up to 1 kGy/h. We developed a radiation resistant optical fiber consisting of a 1000 ppm hydroxyl doped pure silica core and 4 % fluorine doped pure silica cladding. We attempted to apply the optical fiber to remote imaging technique by means of fiberscope. The number of core image fibers was increased from 2000 to 22000 for practical use. The transmissive rate of infrared images was not affected after irradiation of 1 MGy. No change in the spatial resolution of the view scope by means of image fiber was noted between pre- and post-irradiation. We confirmed the applicability of the probing system, which consists of a view scope using radiation-resistant optical fibers.
Sato, Satoshi*; Konno, Chikara; Nakashima, Hiroshi; Shionaga, Ryosuke*; Nose, Hiroyuki*; Ito, Yuji*; Hashimoto, Hirohide*
Journal of Nuclear Science and Technology, 55(4), p.410 - 417, 2018/04
In order to enhance the neutron shielding performance, we developed concrete with boron of more than 10 wt%. We performed a neutron shielding experiment using the mockup of the newly developed boron-loaded concrete and DT neutrons at FNS in JAEA, and measured the reaction rates of the Nb(n,2n)Nb and Au(n,)Au reactions in the mockup. The calculations were conducted by using MCNP-5.14 and FENDL-2.1. The calculation results agreed well with the measured ones, and we confirmed that the accuracy was very good on the atomic composition data of the boron-loaded concrete and their nuclear data. In addition, we calculated effective dose rates and reaction rates of the Co(n,)Co and Eu(n,)Eu reactions in the boron-loaded concrete and other concretes. It is concluded that the boron-loaded concrete has much better shielding performance for DT neutrons than other concretes.
Yamamoto, Takahiro; Ito, Chikara; Maeda, Shigetaka; Ito, Hideaki; Sekine, Takashi
JAEA-Technology 2017-036, 41 Pages, 2018/02
In the experimental fast reactor Joyo, the damaged upper core structure (UCS) was retrieved into the cask in May 2014 The dose rate on UCS surface was quite high due to the activation for over 30 years operation. In order to attain the optimum safety design, manufacture and operation of equipment for UCS replacement, the method to evaluate UCS surface dose rate was developed on the basis of C/E obtained by the in-vessel dose rate measurement in Joyo. In order to verify the evaluation method, the axial gamma-ray distribution measurement on the surface of the cask, which contained UCS, was conducted using a plastic scintillating optical fiber (PSF) detector. This paper describes the comparison results between calculation and measurement as follows. (1) The measured axial gamma-ray distribution on the cask surface had a peak on proper location with considering the cask shielding structure and agree well with the calculated distribution. (2) The C/E of axial gamma-ray distribution on the cask surface was ranged from 1.1 to 1.7. It was confirmed that the calculation for UCS replacement equipment design had a margin conservatively. Then, the results showed that the developed evaluation method for UCS replacement equipment design was sufficiently reliable.
Obana, Tetsuhiro*; Murakami, Haruyuki; Takahata, Kazuya*; Hamaguchi, Shinji*; Chikaraishi, Hirotaka*; Mito, Toshiyuki*; Imagawa, Shinsaku*; Kizu, Kaname; Natsume, Kyohei; Yoshida, Kiyoshi
Physica C, 518, p.96 - 100, 2015/11
Iwata, Yoshihiro; Sekiya, Hiroyuki*; Ito, Chikara
Nuclear Instruments and Methods in Physics Research A, 797, p.64 - 69, 2015/10
An ultrasensitive resonance ionization mass spectrometer that can be applied to evaluate krypton (Kr) contamination in xenon (Xe) dark matter detectors has been developed for measuring Kr at the parts-per-trillion (ppt) or sub-ppt level in Xe. The gas sample is introduced without any condensation into a time-of-flight mass spectrometer through a pulsed supersonic valve. Using a nanosecond pulsed laser at 212.6 nm, Kr atoms in the sample are resonantly ionized along with other Kr isotopes. Kr ions are then mass separated and detected by the mass spectrometer in order to measure the Kr impurity concentration. With our current setup, approximately 0.4 ppt of Kr impurities contained in pure argon (Ar) gas are detectable with a measurement time of 1,000 s. Although Kr detection sensitivity in Xe is expected to be approximately half of that in Ar, our spectrometer can evaluate Kr contamination in Xe to the sub-ppt level.
Murakami, Haruyuki; Kizu, Kaname; Ichige, Toshikatsu; Furukawa, Masato; Natsume, Kyohei; Tsuchiya, Katsuhiko; Kamiya, Koji; Koide, Yoshihiko; Yoshida, Kiyoshi; Obana, Tetsuhiro*; et al.
IEEE Transactions on Applied Superconductivity, 25(3), p.4201305_1 - 4201305_5, 2015/06
JT-60U magnet system will be upgraded to the superconducting coils in the JT-60SA programme of the Broader Approach activities. Terminal joint of Central Solenoid (CS) is wrap type NbSn-NbTi joint used for connecting CS (NbSn) and current feeder (NbTi). The terminal joints are placed at the top and the bottom of the CS systems. CS modules located at middle position of CS system need the lead extension from the modules to the terminal joint. The joint resistance measurement of terminal joint was performed in the test facility of National Institute for Fusion Science. The joint resistance was evaluated by the operating current and the voltage between both ends of the terminal joint part. Test results met the requirement of JT-60SA magnet system. The structural analysis of the lead extension and its support structure was conducted to confirm the support design. In this paper, the results of resistance test of joint and the structural analysis results of lead extension are reported.
Takamatsu, Kuniyoshi; Takegami, Hiroaki; Ito, Chikara; Suzuki, Keiichi*; Onuma, Hiroshi*; Hino, Ryutaro; Okumura, Tadahiko*
Annals of Nuclear Energy, 78, p.166 - 175, 2015/04
In our study, we focused on a nondestructive inspection method by which cosmic-ray muons could be used to observe the internal reactor from outside the RPV and the CV. We conducted an observation test on the HTTR to evaluate the applicability of the method to the internal visualization of a reactor. We also analytically evaluated the resolution of existing muon telescopes to assess their suitability for the HTTR observation, and were able to detect the major structures of the HTTR based on the distribution of the surface densities calculated from the coincidences measured by the telescopes. Our findings suggested that existing muon telescopes could be used for muon observation of the internal reactor from outside the RPV and CV.
Obana, Tetsuhiro*; Takahata, Kazuya*; Hamaguchi, Shinji*; Chikaraishi, Hirotaka*; Mito, Toshiyuki*; Imagawa, Shinsaku*; Kizu, Kaname; Murakami, Haruyuki; Natsume, Kyohei; Yoshida, Kiyoshi
Fusion Engineering and Design, 90, p.55 - 61, 2015/01
In the cold test of the JT-60SA CS model coil made by NbSn CIC conductor, magnetic fields were measured using Hall sensors. While holding coil current of 20 kA, the magnetic fields were varying slightly with several long time constants. The range of the time constant was from 17 sec to 571 sec, which was much longer than the time constant derived from the measurement using the short straight sample. To validate the measurements, the magnetic fields of the model coil were calculated using the calculation model representing the positions of NbSn strands inside the CIC conductor. The calculations were in good agreement with the measurements. Consequently, the validity of magnetic field measurements was confirmed.
Oguri, Shugo*; Kuroda, Yasuhiro*; Kato, Yo*; Nakata, Ryoko*; Inoue, Yoshizumi*; Ito, Chikara; Minowa, Makoto*
Nuclear Instruments and Methods in Physics Research A, 757, p.33 - 39, 2014/09
We developed a segmented reactor antineutrino detector made of plastic scintillators for the nuclear safeguard application and demonstrate almost unmanned field operation at a commercial power plant reactor. We observed the difference of the reactor antineutrino flux with the reactor ON and OFF above the ground outside the reactor building.
Saeki, Morihisa; Iwanade, Akio; Ito, Chikara; Wakaida, Ikuo; Thornton, B.*; Sakka, Tetsuo*; Oba, Hironori
Journal of Nuclear Science and Technology, 51(7-8), p.930 - 938, 2014/07
To inspect post-accident nuclear core reactor of the TEPCO Fukushima Daiichi Nuclear Power Plant, a transportable fiber-coupled LIBS instrument was developed. The developed LIBS instrument was designed to analyze the underwater sample in high-radiation field by single-pulse breakdown with gas flow or double-pulse breakdown. To check the feasibility of the assembled fiber-coupled LIBS instrument to the inspection inside Fukushima Daiichi Nuclear Power Plant, we investigated (1) influence of radiation dose on optical transmittance of the laser delivery fiber, (2) survey of the LIBS techniques to analyze the underwater sample and (3) candidates of emission lines for analysis of the debris. By employing the selected emission lines, we demonstrated that the developed LIBS instrument can analyze the simulated debris underwater by the single-pulse breakdown with the gas flow with high signal to noise ratio.
Ito, Chikara; Naito, Hiroyuki; Nishimura, Akihiko; Oba, Hironori; Wakaida, Ikuo; Sugiyama, Akira; Chiyatani, Keiji
Journal of Nuclear Science and Technology, 51(7-8), p.944 - 950, 2014/07
In the Fukushima Daiichi Nuclear Power Station, it is necessary to survey the locations and conditions of fuel debris inside reactor pressure vessels or primary containment vessels under water and radiation environment in preparation for removing fuel debris. An optical fiber is well known for features such as signal transmission, light weight, superior insulation performance, water resistance and electromagnetic noise resistance. These features allow the optical fiber to simplify the instrumentation systems for in-vessel inspection, as long as provide that the optical fiber can be used under high radiation dose environment. The radiation resistance of an optical fiber was improved by increasing the amount of hydroxyl up to 1000 ppm in pure silica fiber. The improved optical fibers were irradiated with -ray up to 1 10 Gy using a Co source. They indicated a large peak around 600 nm and a peak tail from ultraviolet region, but no large absorption in infrared region except a hydroxyl absorption peak of 945 nm. We have confirmed that the optical fiber containing 1000 ppm hydroxyl has enough radiation resistance for radiation induced transmission losses and the infrared imaging is effective for observation under high radiation doses.
Ito, Chikara; Naito, Hiroyuki; Oba, Hironori; Saeki, Morihisa; Ito, Keisuke; Ishikawa, Takashi; Nishimura, Akihiko; Wakaida, Ikuo; Sekine, Takashi
Proceedings of 22nd International Conference on Nuclear Engineering (ICONE-22) (DVD-ROM), 7 Pages, 2014/07
A high-radiation resistant optical fiber has been developed in order to investigate the interiors of the reactor pressure vessels and the primary containment vessels of the Fukushima Daiichi Nuclear Power Station. The radiation resistance of an optical fiber was improved by increasing the amount of hydroxyl up to 1000 ppm in pure silica fiber. The improved image fiber consists of common cladding and a large number of fiber cores made from pure silica that contains 1000 ppm hydroxyl. The transmissive rate of an infrared image was not affected after the irradiation of 1 MGy. We have developed the fiber-coupled LIBS system to detect plasma emission efficiently in near-infrared region. In addition, we have performed a ray dose rate measurement using an optical fiber of which scintillator is attached to the tip. As a result, the concept of applicability of a probing system using the high-radiation resistant optical fibers has been confirmed.
Murakami, Haruyuki; Kizu, Kaname; Tsuchiya, Katsuhiko; Koide, Yoshihiko; Yoshida, Kiyoshi; Obana, Tetsuhiro*; Takahata, Kazuya*; Hamaguchi, Shinji*; Chikaraishi, Hirotaka*; Natsume, Kyohei*; et al.
IEEE Transactions on Applied Superconductivity, 24(3), p.4200205_1 - 4200205_5, 2014/06
Central Solenoid (CS) of JT-60SA are designed with the NbSn cable in conduit conductor. CS model coil (CSMC) was manufactured by using the real manufacturing jigs and procedure to validate the CS manufacturing processes before starting mass production. The dimensions of the CSMC are the same as real quad-pancake. The cold test of the CSMC was performed and the test results satisfied the design requirements. These results indicate that the manufacturing processes of the JT-60SA CS has been established. In this paper, the development and the validation of the CS manufacturing processes are described.
Iwata, Yoshihiro; Ito, Chikara; Harano, Hideki*; Iguchi, Tetsuo*
Journal of Nuclear Science and Technology, 51(4), p.465 - 475, 2014/04
To prevent a fuel failure event from becoming a serious radiation accident, sodium-cooled fast reactors are equipped with a system for failed fuel detection and location (FFDL). The FFDL instrument employed in the prototype fast breeder reactor Monju is based on the gas tagging method, in which precise and accurate measurements of krypton and xenon isotope ratios (Kr/Kr, Kr/Kr and Xe/Xe) must be performed in a short time. Burnup measurements also contribute to accurate determination of Kr/Kr. We have developed a highly sensitive resonance ionization mass spectrometer for the isotopic analyses, which uses resonance ionization of Kr and Xe atoms by a pulsed laser at wavelengths of 216.7 nm and 249.6 nm, respectively. In evaluating the performance of our spectrometer, we find that systematic errors caused by isotope shifts can be reduced to negligible levels, and that statistical errors of 3% at a nuclide concentration of 7 ppt can be achieved with a single measurement time of about 40 minutes for each Kr and Xe isotope ratio. This means that, within one hour, about 200 fuel assemblies can be individually identified with a probability of 99%, verifying the applicability of our spectrometer to the FFDL system of fast reactors.
Takegami, Hiroaki; Takamatsu, Kuniyoshi; Ito, Chikara; Hino, Ryutaro; Suzuki, Keiichi*; Onuma, Hiroshi*; Okumura, Tadahiko*
Nippon Genshiryoku Gakkai Wabun Rombunshi, 13(1), p.7 - 16, 2014/03
One of the important problems for controlling of the Fukushima Daiichi Nuclear Power Plant is removing of fuel debris. As this preparation, the nondestructive inspection method for grasping the position of fuel debris is required. Therefore, we focused on a nondestructive inspection method using cosmic-ray muons. In this study, the applicability of this method for internal visualization of reactor was confirmed by preliminary test of internal visualization of High Temperature Engineering Test Reactor (HTTR). By using cosmic-ray muons, major components in the HTTR, such as concrete wall and reactor core, can be observed from the outside of a containment vessel. From the results, it appears that the inspection method with muons is a candidate method for searching the fuel debris in a reactor. Based on the results, we also proposed some improvements of this system for inspection at the Fukushima Daiichi Nuclear Power Station.
Takasaki, Koji; Yasumune, Takashi; Onishi, Takashi; Nakamura, Keisuke; Ishimi, Akihiro; Ito, Chikara; Osaka, Masahiko; Ono, Masashi*; Hatakeyama, Shuichi*; Takahashi, Hiroyuki*; et al.
JAEA-Research 2013-043, 33 Pages, 2014/01
In the Fukushima Daiichi Nuclear Power Plant, it is assumed that the core fuels melted partially or wholly, and the normal technique of accounting for a fuel assembly is not applicable. Therefore, it is necessary to develop the transparent and rational technique of accounting in the process of collection and storage of fuel debris. In this research, an application of the superconducting phase Transition Edge Sensor microcalorimeter (TES microcalorimeter) is studied for the accounting of nuclear materials in the fuel debris. It is expected that the detailed information of nuclear materials and fission products in fuel debris is obtained by using a high-resolution characteristic of TES microcalorimeter. In this report, the principle of TES microcalorimeter, the measurement experiment using TES in JAEA, and the simulated calculation using the EGS5 code system are summarized.
Takegami, Hiroaki; Terada, Atsuhiko; Noguchi, Hiroki; Kamiji, Yu; Ono, Masato; Takamatsu, Kuniyoshi; Ito, Chikara; Hino, Ryutaro; Suzuki, Keiichi*; Onuma, Hiroshi*; et al.
JAEA-Research 2013-032, 25 Pages, 2013/12
We focused on a non-destructive inspection method using cosmic-ray muons as a candidate method for observation of internal the reactor from the outside of a reactor building. In this study, the applicability of this method for the reactor investigation was confirmed by a preliminary examination with High Temperature Engineering Test Reactor (HTTR). From the results of this examination, it appears that high density structures, such as the core and concrete walls, were able to observe by using muon telescope with coincidence method from the outside of the pressure vessel. Furthermore, we proposed some improvements of this muon inspection system for on-site investigation at the Fukushima Daiichi NPS.
Kamiji, Yu; Akai, Naoki*; Ozaki, Koichi*; Ito, Chikara
JAEA-Research 2013-018, 18 Pages, 2013/11
Preliminary examination of image processing was conducted using existing reactor inside images through a fiber scope in order to confirm applicability of 3D-mapping in a reactor as a part of development of 3D-measurement technology. The Upright SURF (Speeded Up Robust Features) was used to find corresponding points between two captured images. In case of images showing many similar textures or lack of texture, it was difficult to find corresponding points using SURF only. By coupling with a canny algorithm to detect edges of the inside structure, it was found that 3D-structure could be measured for rectilinear objects. Also, image acquisition model was examined in order to clarify camera specs.
Ito, Chikara; Ito, Keisuke; Ishikawa, Takashi; Yoshida, Akihiro; Sanada, Yukihisa; Torii, Tatsuo; Notomi, Akihiro*; Wakabayashi, Genichiro*; Miyazaki, Nobuyuki*
Hoshasen, 39(1), p.7 - 11, 2013/09
no abstracts in English