Suzuki, Yumi*; Nakano, Hiroko; Suzuki, Yoshitaka; Ishida, Takuya; Shibata, Akira; Kato, Yoshiaki; Kawamata, Kazuo; Tsuchiya, Kunihiko
JAEA-Technology 2015-031, 58 Pages, 2015/11
Technetium-99m (Tc) is one of the most commonly used radioisotopes in the field of nuclear medicine. In the Japan Atomic Energy Agency (JAEA), the research and development (R&D) have been carried out for production of molybdenum-99 (Mo) by (n, ) method, a parent nuclide of Tc, with the Japan Material Testing Reactor (JMTR). On the other hand, the new project as "Domestic Production of Medical Radioisotope (Technetium preparation) in Japan" was adopted in the Tsukuba International Strategic Zone on October, 2013 and the demonstration tests will be planned for the domestic production of Mo/Tc with the JMTR. Thus, new facilities and analysis devices were equipped in the JMTR Hot Laboratory in 2014 as the part of this project. As the part of the analytical device equipment, the -TLC analyzer and the radiation detector connected with the High Performance Liquid Chromatography (HPLC) were installed for quality inspection of the Mo/Tc solution and the extracted Tc solution in the JMTR Hot Laboratory. The performance tests of these devices such as detection sensitivity, resolution, linearity and selectivity of energy range were carried out with Cs and Eu as alternative radionuclides of Mo and Tc, respectively. In the results, bright prospects were obtained concerning the quality inspection of the Mo/Tc and Tc solutions using these devices. This report describes the results of those performance tests.
Takechi, Manabu; Matsunaga, Go; Sakurai, Shinji; Sasajima, Tadayuki; Yagyu, Junichi; Hoshi, Ryo*; Kawamata, Yoichi; Kurihara, Kenichi; JT-60SA Team; Nishikawa, T.*; et al.
Fusion Engineering and Design, 96-97, p.985 - 988, 2015/10
Shibata, Akira; Kato, Yoshiaki; Oishi, Makoto; Taguchi, Taketoshi; Ito, Masayasu; Yonekawa, Minoru; Kawamata, Kazuo
KAERI/GP-418/2015, p.151 - 165, 2015/05
The JMTR stopped its operation in 2006 for refurbishment. The reactor facilities have been refurbished from 2007. After refurbishment, JMTR Hot laboratory is expected to perform various post irradiation examinations. In this report, installations of experimental apparatuses and recent experimental method are introduced. (1) A nano-indenter with radius spherical indenter. Inverse analysis using FEM could presume material constants from load-depth curve of indentation. Mechanical properties of oxide layer of zirconium alloy and irradiated stainless steel will be analyzed. (2) Transmission Electron Microscope (TEM). TEM is capable of imaging at a significantly higher resolution than light microscopes or normal SEM. JAEA installed a TEM apparatus (JEOL JEM-2800) in JMTR Hot laboratory. The maximum magnification is 150,000,000 times. It can be operated from a remote location using a computer network. This contributes to the convenience of remote researchers and reducing the amount of exposure.
Kondo, Yasuhiro; Morishita, Takatoshi; Yamazaki, Saishun; Hori, Toshihiko; Sawabe, Yuki; Chishiro, Etsuji; Fukuda, Shimpei; Hasegawa, Kazuo; Hirano, Koichiro; Kikuzawa, Nobuhiro; et al.
Physical Review Special Topics; Accelerators and Beams, 17(12), p.120101_1 - 120101_8, 2014/12
We performed a beam test of a new radio frequency quadrupole linac (RFQ III) for the beam current upgrade of the Japan Proton Accelerator Research Complex. First, the conditioning of RFQ III was conducted, and after 20 h of conditioning, RFQ III became very stable with a nominal peak power and duty factor of 400 kW and 1.5%, respectively. An off-line beam test was subsequently conducted before installation in the accelerator tunnel. The transmission, transverse emittance, and energy spread of the 50-mA negative hydrogen beam from RFQ III were measured and compared with simulation results. The experiment and simulation results showed good agreement; therefore, we conclude that the performance of RFQ III conforms to its design.
Watahiki, Shunsuke; Hanakawa, Hiroki; Imaizumi, Tomomi; Nagata, Hiroshi; Ide, Hiroshi; Komukai, Bunsaku; Kimura, Nobuaki; Miyauchi, Masaru; Ito, Masayasu; Nishikata, Kaori; et al.
JAEA-Technology 2013-021, 43 Pages, 2013/07
The number of research reactors in the world is decreasing because of their aging. On the other hand, the necessity of research reactor, which is used for human resources development, progress of the science and technology, industrial use and safety research is increasing for the countries which are planning to introduce the nuclear power plants. From above background, the Neutron Irradiation and Testing Reactor Center began to discuss a basic concept of Multipurpose Compact Research Reactor (MCRR) for education and training, etc., on 2010 to 2012. This activity is also expected to contribute to design tool improvement and human resource development in the center. In 2011, design study of reactor core, irradiation facilities with high versatility and practicality, and hot laboratory equipment for the production of Mo-99 was carried out. As the result of design study of reactor core, subcriticality and operation time of the reactor in consideration of an irradiation capsule, and about the transient response of the reactor to the reactivity disturbance during automatic control operation, it was possible to do automatic operation of MCRR, was confirmed. As the result of design study of irradiation facilities, it was confirmed that the implementation of an efficient mass production radioisotope Mo-99 can be expected. As the result of design study with hot laboratory facilities, Mo-99 production, RI export devised considered cell and facilities for exporting the specimens quickly was designed.
Kondo, Yasuhiro; Morishita, Takatoshi; Hasegawa, Kazuo; Chishiro, Etsuji; Hirano, Koichiro; Hori, Toshihiko; Oguri, Hidetomo; Sato, Fumiaki; Shinozaki, Shinichi; Sugimura, Takashi*; et al.
Physical Review Special Topics; Accelerators and Beams, 16(4), p.040102_1 - 040102_8, 2013/04
We performed a high-power test of a new radio-frequency quadrupole (RFQ II) for the J-PARC linac. RFQ II was developed as a spare RFQ because the operating J-PARC RFQ has suffered from a sparking problem. First, the conditioning of RFQ II was carried out; after 50 h of conditioning, RFQ II became very stable with a nominal power and duty of 330 kW and 3%, respectively. Next the thermal properties were measured because the resonant frequency of RFQ II is tuned by changing the temperature of the cooling water. The frequency response was measured and compared to finite-element method simulation results, confirming that the simple two-dimensional model reproduces the experimental data well. The differences in the field distribution with changes in the rf loading and the cooling-water temperature were also measured, and no serious field distortion was observed. Therefore, we conclude that RFQ II can perform well as a high-power rf cavity.
Taguchi, Taketoshi; Yonekawa, Minoru; Kato, Yoshiaki; Kurosawa, Makoto; Nishikata, Kaori; Ishida, Takuya; Kawamata, Kazuo
UTNL-R-0483, p.10_5_1 - 10_5_13, 2013/03
JMTR focus on the activation method. By carrying out the preliminary tests using irradiation facilities existing, and verification tests using the irradiation facility that has developed in the cutting-edge research and development strategic strengthening business, as irradiation tests towards the production of Mo, we have been conducting research and development that can contribute to supply about 25% for Mo demand in Japan and the stable supply of radiopharmaceutical. This report describes a summary of the status of the preliminary tests for the production of Mo: Maintenance of test equipment in the facility in JMTR Hot Laboratory in preparation for research and development for the production of Mo in JMTR and using MoO pellet irradiated at Kyoto University Research Reactor Institute (KUR).
Kondo, Yasuhiro; Morishita, Takatoshi; Hasegawa, Kazuo; Kawamata, Hiroshi*; Sugimura, Takashi*; Naito, Fujio*
Proceedings of 26th International Linear Accelerator Conference (LINAC 2012) (Internet), p.972 - 974, 2012/09
In this paper, we will report recent topics about J-PARC RFQs. First, the operating RFQ (RFQ I) have been recovered from the long shutdown due to the earthquake. This RFQ have been suffered from breakdown problem since 2008, therefore we have been developing a back-up RFQ (RFQ II). In April 2012, the high power test was successfully performed. It was confirmed to be able to stably operate at nominal power and duty factor (333 kW and 3%). Finally, we are fabricating a new RFQ for the beam-current upgrade of the J-PARC linac (RFQ III). The brazing of the first unit will be done in this September. The status of these RFQs are described.
Shibata, Akira; Nakano, Junichi; Omi, Masao; Kawamata, Kazuo; Nakagawa, Tetsuya; Tsukada, Takashi
Journal of Nuclear Materials, 422(1-3), p.14 - 19, 2012/03
To simulate Irradiation assisted stress corrosion cracking (IASCC) behavior by in-pile experiments, it is necessary to irradiate specimens up to a neutron fluence that is higher than the IASCC threshold fluence. Pre-irradiated specimens must be relocated from pre-irradiation capsules to in-pile capsules. Hence, a remote welding machine has been developed. And the integrity of capsule housing for a long term irradiation was evaluated by tensile tests in air and slow strain rate tests in water. Two type specimens were prepared. Specimens were obtained from the outer tubes of capsule irradiated to 1.0-3.9 10 n/m (E 1 MeV). And specimens were irradiated in a leaky capsule to 0.03-1.0 10 n/m. Elongation more than 15% in tensile test at 423 K was confirmed and no IGSCC fraction was shown in SSRT at 423 K which was estimated as temperature at the outer tubes of the capsule under irradiation.
Kondo, Yasuhiro; Hasegawa, Kazuo; Kawamata, Hiroshi*; Morishita, Takatoshi; Naito, Fujio*
Nuclear Instruments and Methods in Physics Research A, 667, p.5 - 10, 2012/03
We have developed a new on-machine non-contact dimension-measurement system using a laser displacement sensor (LDS) for machining the vane tips of RFQs. The LDS was attached to a milling machine, and the dimensions of a test piece (TP) with a modulated vane tip were measured to evaluate the performance of the system. A longitudinal profile was measured with the LDS and it was compared with the profile measured with a coordinate measuring machine (CMM). A transverse profile was also measured. The required accuracy of dimension measurement was achieved using our system.
Imaizumi, Tomomi; Miyauchi, Masaru; Ito, Masayasu; Watahiki, Shunsuke; Nagata, Hiroshi; Hanakawa, Hiroki; Naka, Michihiro; Kawamata, Kazuo; Yamaura, Takayuki; Ide, Hiroshi; et al.
JAEA-Technology 2011-031, 123 Pages, 2012/01
The number of research reactors in the world is decreasing because of their aging. However, the planning to introduce the nuclear power plants is increasing in Asian countries. In these Asian countries, the key issue is the human resource development for operation and management of nuclear power plants after constructed them, and also the necessity of research reactor, which is used for lifetime extension of LWRs, progress of the science and technology, expansion of industry use, human resources training and so on, is increasing. From above backgrounds, the Neutron Irradiation and Testing Reactor Center began to discuss basic concept of a multipurpose low-power research reactor for education and training, etc. This design study is expected to contribute not only to design tool improvement and human resources development in the Neutron Irradiation and Testing Reactor Center but also to maintain and upgrade the technology on research reactors in nuclear power-related companies. This report treats the activities of the working group from July 2010 to June 2011 on the multipurpose low-power research reactor in the Neutron Irradiation and Testing Reactor Center and nuclear power-related companies.
Nakamura, Kazuo*; Jiang, Y.*; Liu, X.*; Mitarai, Osamu*; Kurihara, Kenichi; Kawamata, Yoichi; Sueoka, Michiharu; Hasegawa, Makoto*; Tokunaga, Kazutoshi*; Zushi, Hideki*; et al.
Fusion Engineering and Design, 86(6-8), p.1080 - 1084, 2011/10
Ito, Masayasu; Kawamata, Kazuo; Tayama, Yoshinobu; Kanazawa, Yoshiharu; Yonekawa, Minoru; Nakagawa, Tetsuya; Omi, Masao; Iwamatsu, Shigemi
JAEA-Technology 2011-022, 44 Pages, 2011/07
Hot laboratory are facilities that execute the post irradiation examination of sample irradiated in material testing reactors etc. The handling of high burn-up fuel is scheduled in the JMTR (Japan Materials Testing Reactor) Hot Laboratory with JMTR re-operate in FY 2011. This report describes evaluation, production and installation of shielding of the hot cells in the JMTR Hot Laboratory.
Ito, Masayasu; Kawamata, Kazuo; Sozawa, Shizuo; Shibata, Akira; Tsuchiya, Kunihiko; Kitsunai, Yuji*; Kodama, Mitsuhiro*
JAEA-Testing 2011-003, 22 Pages, 2011/06
In JMTR, beryllium metal is used for neutron reflector. However, the periodical exchange of the neutron reflector is necessary due to degradation of strength and swelling under neutron irradiation. So it is required to select the material that bears long term from the viewpoint of the utilization rate improvement and the radioactive waste decrease in JMTR. It is important to evaluate the characteristics of beryllium metals precisely. This report describes the preparation method of the sample and observation results by the transmission electron microscope that is being examined as a evaluation method of the beryllium metal.
Ishitsuka, Etsuo; Kitagishi, Shigeru; Aoyama, Masashi; Kawamata, Kazuo; Nagao, Yoshiharu; Ishihara, Masahiro; Kawamura, Hiroshi
Proceedings of 1st Asian Symposium on Material Testing Reactors (ASMTR 2011), p.111 - 115, 2011/02
Tanaka, Kosuke; Kawamata, Kazuo; Yoshimochi, Hiroshi; Sozawa, Shizuo; Onose, Shoji; Niimi, Motoji; Asaka, Takeo
Proceedings of 1st Asian Symposium on Material Testing Reactors (ASMTR 2011), p.71 - 76, 2011/02
Post irradiation examination (PIE) facilities have been operated for about 40 years at the Oarai Research and Development Center of the Japan Atomic Energy Agency to investigate the performance and soundness of irradiated fuels and materials. The JMTR Hot Laboratory (JMTR-HL) was founded in 1971 mainly to examine the objects irradiated in the Japan Material Testing Reactor (JMTR). The Alpha-Gamma Facility (AGF) was constructed as the first laboratory to perform PIE of plutonium-bearing fuels for Japanese fast reactor development programs. This facility started hot operation in 1971 and has performed physical, metallurgical, and chemical examinations of irradiated fuels including uranium plutonium mixed oxide fuels. A renewal plan for the JMTR-HL and AGF is now in progress, associated with re-operation of the JMTR.
Morishita, Takatoshi; Kondo, Yasuhiro; Hasegawa, Kazuo; Naito, Fujio*; Yoshioka, Masakazu*; Matsumoto, Hiroshi*; Hori, Yoichiro*; Kawamata, Hiroshi*; Saito, Yoshio*; Baba, Hiroshi*
Proceedings of 25th International Linear Accelerator Conference (LINAC 2010) (CD-ROM), p.521 - 523, 2010/09
The fabrication of a new RFQ has been started as a backup machine for the J-PARC linac. The RFQ cavity is divided by three unit tanks in the longitudinal direction. The unit tank consists of two major vanes and two minor vanes, those are brazed together. A one-step vacuum brazing of a unit tank was adopted to unite these four vanes together with the flanges and ports. At the first tank brazing, the vacuum leak has been occurred due to the non-uniform temperature rise during the heating. Repair of this leakage and the results of the improved brazing of the second tank are reported.
Morishita, Takatoshi; Kondo, Yasuhiro; Hasegawa, Kazuo; Kawamata, Hiroshi*; Naito, Fujio*; Yoshioka, Masakazu*; Matsumoto, Hiroshi*; Hori, Yoichiro*; Saito, Yoshio*; Baba, Hiroshi*
Proceedings of 25th International Linear Accelerator Conference (LINAC 2010) (CD-ROM), p.518 - 520, 2010/09
The fabrication of a new RFQ has been started in the J-PARC linac. The RFQ cavity is divided by three unit tanks in the longitudinal direction. The unit tank consists of two major vanes and two minor vanes, those will be brazed together. To reduce the costs and periods to develop the special formed bite for the modulation machining, a numerical controlled machining using a conventional ball-end-mill was adopted for the vane modulation cutting instead of the wheel shape cutter. The dimension accuracy was confirmed by cutting test pieces. Moreover, the obtained surface seems smooth enough for the operation. Results of the ball-end-mill machining for the vanes are described.
Morishita, Takatoshi; Kondo, Yasuhiro; Hasegawa, Kazuo; Naito, Fujio*; Yoshioka, Masakazu*; Matsumoto, Hiroshi*; Hori, Yoichiro*; Kawamata, Hiroshi*; Saito, Yoshio*; Baba, Hiroshi*
Proceedings of 1st International Particle Accelerator Conference (IPAC '10) (Internet), p.783 - 785, 2010/05
The J-PARC RFQ (length 3.1 m, 4-vane type, 324 MHz) accelerates a negative hydrogen beam from 0.05 MeV to 3 MeV toward the following DTL. The stability of the operating RFQ decreased for a few months at the end of 2008, then, we started a preparation of a new RFQ as a backup machine in the case of RFQ problem. The beam dynamics design of the new RFQ is the same as the current cavity for a quick resumption of operation, however, the engineering and RF designs are changed to improve stability. The processes of the vane machining and the surface treatments have been carefully considered to reduce the discharge problem. The vacuum brazing technique has been chosen for vane jointing. The design of the new RFQ and the fabrication progress is descried in this proceeding.
Ito, Masayasu; Kawamata, Kazuo; Ishitsuka, Etsuo
JAEA-Testing 2009-007, 16 Pages, 2010/03
Surface state of material is measured by the surface displacement measurement device with laser displacement sensor. It is important to know the characteristic of a device that considers environment to use this device in the cell. Therefore, this report described about evaluation of characteristics of the device by laser method.