Naoe, Takashi; Kogawa, Hiroyuki; Wakui, Takashi; Teshigawara, Makoto; Haga, Katsuhiro; Futakawa, Masatoshi
Nuclear Instruments and Methods in Physics Research A, 982, p.164566_1 - 164566_6, 2020/12
A liquid mercury target for the spallation neutron source is installed in the J-PARC. The liquid mercury is enclosed with the multi-walled stainless steel vessel. At the time of highly intense proton beams hits the target at a repetition rate of 25 Hz, pressure waves, that causes cavitation erosion, are generated owing the rapidly thermal expansion of mercury. We have installed the target diagnostic system consisting of a laser Doppler vibrometer (LDV) and a dynamic microphone to remotely investigate the structural integrity of the target under high-radiation environment. In this study, aiming to understand correlation between the acoustic vibration and the operation conditions such as the proton beam power and beam profile, proton beam induced acoustic vibration was measured by parametrically changing the target operation conditions. The result showed that the sound is well correlated with the operation conditions.
Sakai, Kenji; Oi, Motoki; Teshigawara, Makoto; Naoe, Takashi; Haga, Katsuhiro; Watanabe, Akihiko*
Journal of Neutron Research, 22(2-3), p.337 - 343, 2020/10
For operating a spallation neutron source and a muon target safely and efficiently, a general control system (GCS) operates within Materials and Life Science Experimental Facility (MLF). GCS administers operation and interlock processes of many instruments under various operation status. Since the first beam injection in 2008, it has operated stably without any serious troubles for more than ten years. GCS has a data storage server storing operational data on status around target stations. It has functioned well to detect and investigate unusual situations by checking data in this server. For continuing stable operation of MLF in future, however, introduction of abnormality sign determination system (ASDS) will be necessary for picking up potential abnormalities of target stations caused by radiation damages, time-related deterioration and so on. It will judge abnormalities from slight state transitions of target stations based on analysis with various operational data throughout proton beams, target stations, and secondary beams during long-term operations. This report mentions present status of GCS, conceptual design of ASDS, and installation of an integral data storage server which can deal with various data for ASDS integrally.
Kogawa, Hiroyuki; Wakui, Takashi; Naoe, Takashi; Haga, Katsuhiro; Takada, Hiroshi; Futakawa, Masatoshi
Journal of Nuclear Science and Technology, 57(5), p.487 - 494, 2020/05
Neutron flux per pulse reached world record at neutron source in the J-PARC. In the J-PARC, mercury target system is used as a spallation neutron source. A target vessel has the multi-walled protection system that comprises a mercury vessel enclosed with a double-walled water shroud. This is to prevent the leakage of the mercury outside the mercury vessel. The multi-walled structure needed to be complicated with a lot of welding lines. However, during the operation, we faced an unscheduled shutdown due to water leakage to the intermediate layer between the mercury vessel and water shroud. An investigation on the cause of the leakage was carried out. It is deduced that the leakage path was formed due to the crack propagation from welding defects that is caused by the complicated multi-walled structure. The crack propagation is attributed to the repeated stress by pressure waves generated in the mercury target. Based on the investigation results, the design was improved to remove the welding line on the complicated structure and to realize the stable operation with 1 MW proton beam.
Naoe, Takashi; Kinoshita, Hidetaka; Kogawa, Hiroyuki; Wakui, Takashi; Wakai, Eiichi; Haga, Katsuhiro; Takada, Hiroshi
JPS Conference Proceedings (Internet), 28, p.081004_1 - 081004_6, 2020/02
The beam window of the mercury target vessel in J-PARC is severely damaged by the cavitation. The cavitation damage is a crucial factor to limit lifetime of the target because it increases with the beam power. Therefore, mitigating cavitation damage is an important issue to operate the target stably for long time at 1 MW. At J-PARC, to mitigate the cavitation damage: gas microbubbles are injected into mercury for suppressing pressure waves, and double-walled structure with a narrow channel of 2 mm in width to form high-speed mercury flow (4m/s) has been adopted. After operation, the beam window was cut to inspect the effect of the cavitation damage mitigation on inner wall. We optimized cutting conditions through the cold cutting tests, succeeding in cutting the target No.2 (without damage mitigation technologies) smoothly in 2017, and target No.8 with damage mitigation technologies. In the workshop, progress of cavitation damage observation for the target vessel will be presented.
Naoe, Takashi; Harjo, S.; Kawasaki, Takuro; Xiong, Z.*; Futakawa, Masatoshi
JPS Conference Proceedings (Internet), 28, p.061009_1 - 061009_6, 2020/02
At the J-PARC, a mercury target vessel made of 316L SS suffers proton and neutron radiation environment. The target vessel also suffers cyclic impact stress caused by the proton beam-induced pressure waves. The vessel suffers higher than 4.510 cyclic loading during the expected service life of 5000 h. We have investigated fatigue strength 316L SS up to gigacycle in the previous studies. The cyclic hardening and softening behavior were observed. In this study, to evaluate the cyclic hardening/softening behavior, the dislocation densities of specimens were measured using the neutron diffraction method at the MLF BL-19. The result showed that the dislocation density of a 316L SS was increased with increasing the number of loading cycles. By contrast, in the case of cold-rolled 316L SS, annihilation and re-accumulation of dislocation by cyclic loading were observed. In the workshop, result of neutron diffraction measurement will be introduced with the progress of fatigue test.
Wakui, Takashi; Wakai, Eiichi; Kogawa, Hiroyuki; Naoe, Takashi; Hanano, Kohei; Haga, Katsuhiro; Takada, Hiroshi; Shimada, Tsubasa*; Kanomata, Kenichi*
JPS Conference Proceedings (Internet), 28, p.081002_1 - 081002_6, 2020/02
A mercury target vessel of J-PRAC is designed with a triple-walled structure consisting of the mercury vessel and a double-walled water shroud with internal and external vessels. During the beam operation at 500 kW in 2015, small water leakages from a water shroud of the mercury target vessel occurred twice. Design, fabrication and inspection processes were improved based on the lessons learned from the target failures. The total length of welding lines at the front of the mercury target vessel decreases drastically to approximately 55% by adopting monolithic structure cut out from a block of stainless steel by the wire-electrical discharge machining. Thorough testing of welds by radiographic testing and ultrasonic testing was conducted. The fabrication of the mercury target vessel #8 was finished on September 2017 and the beam operation using it started. Stable beam operation at 500 kW has been achieved and it could experience the maximum beam power of 1 MW during a beam test.
Naoe, Takashi; Kogawa, Hiroyuki; Tanaka, Nobuatsu*; Futakawa, Masatoshi
Advanced Experimental Mechanics, 4, p.17 - 21, 2019/08
We have introduced the following two techniques to mitigate the pressure wave-induced cavitation damage in the mercury target. One is the gas microbubble injection into the flowing mercury, and the other is the double-walled structure with a narrow gap channel at the proton beam entrance portion of the mercury vessel. The latter is expected to mitigate the cavitation damage due to the high-speed liquid flow ( 4 m/s) and the narrow gap boundary (2 mm). To quantitatively investigate the effect of double-walled structure on cavitation damage, cavitation damage tests were conducted by parametrically changing mercury flow velocity and gap width of the channel wall. The results showed that the damage evaluated as a surface roughness was reduced by increasing the flow velocity. By contrast, the effect of gap width on cavitation damage was hardly observed under flowing conditions.
Kawamura, Shunsuke; Naoe, Takashi; Ikeda, Tsubasa*; Tanaka, Nobuatsu*; Futakawa, Masatoshi
Advanced Experimental Mechanics, 4, p.33 - 37, 2019/08
A mercury enclosure vessel made of stainless steel is used as a spallation target in the pulsed spallation neutron source at J-PARC. It is severely damaged by the cavitation induced with pressure waves in association with the pulsed proton beam injection. A double-walled structure with a narrow mercury channel was adopted in the front end of the target vessel to reduce the cavitation damage. It has been experimentally demonstrated that the cavitation damage could be mitigated in the narrow channel but its mechanism has been unclarified yet. In this study, we investigated the cavitation from growing to collapsing through visualizing the spark-induced cavitation bubbles under flow field using a high-speed video camera. Furthermore, we measured the wall vibration due to the cavitation bubble collapse with changing flow velocity parametrically. It was found that the microjet collided perpendicular to the wall in the stagnant flow condition while it collided with an inclined angle from the perpendicular direction, suggesting that the collision pressure on the wall was reduced by flowing.
Wakui, Takashi; Ishii, Hideaki*; Naoe, Takashi; Kogawa, Hiroyuki; Haga, Katsuhiro; Wakai, Eiichi; Takada, Hiroshi; Futakawa, Masatoshi
Materials Transactions, 60(6), p.1026 - 1033, 2019/06
The mercury target has large size as 184.108.40.206 m. In view of reducing the amount of wastes, we studied the structure so that the fore part could be separated. The flange is required to have high seal performance less than 110 Pa m/s. Invar with low thermal expansion is a candidate. Due to its low stiffness, however, the flange may deform when it is fastened by bolts. Practically invar is reinforced with stainless steel where all interface between them has to be bonded completely with the HIP bonding. In this study, we made specimens at four temperatures and conducted tensile tests. The specimen bonded at 973 K had little diffusion layer, and so fractured at the interface. The tensile strength reduced with increasing the temperature, and the reduced amount was about 10% at 1473 K. The analyzed residual stresses near the interface increased by 50% at maximum. Then, we concluded that the optimum temperature was 1173 K.
Naoe, Takashi; Teshigawara, Makoto; Futakawa, Masatoshi; Mizutani, Haruki; Muramatsu, Toshiharu; Yamada, Tomonori; Ushitsuka, Yuji*; Tanaka, Nobuatsu*; Yamasaki, Kazuhiko*
Proceedings of 8th International Congress on Laser Advanced Materials Processing (LAMP 2019) (Internet), 5 Pages, 2019/05
Laser cutting is one of the options in the disposal of radio-active waste, such as spallation neutron target vessel in J-PARC, etc. Due to unique characteristic of laser, such as non-contact system, it is more easily to provide remote-controlled system in comparison with conventional one, such as mechanical cutting machine, etc. However, a demerit of laser cutting is the sputter and fume caused by laser cutting, resulting in contamination with radio-active materials its surroundings. Recently it was developed that the spatter suppression technique by controlling laser beam profile in laser welding process. In order to apply this suppression technique to laser cutting, first of all, we attempted to observe the phenomenon at melting area during laser cutting using a high-speed video camera in order to make the physical model. The result showed that the appearance of fume and sputter were independently confirmed in the time evolution.
Xiong, Z.*; Naoe, Takashi; Futakawa, Masatoshi
Metals, 9(4), p.412_1 - 412_11, 2019/04
The effect of surface defects on the very high cycle fatigue (VHCF) behavior were investigated on the solution annealed (SA) and cold-rolled (CW) 316L. Surface defects were artificially created using indentation. VHCF test was conducted using an ultrasonic fatigue method. The results showed that the fatigue crack initiation was independent of the indent with the applied range of depth in this research. Furthermore, the critical depth of the indent was evaluated based on an empirical formula. In the case of SA, the VHCF strength was not affected when the indent depth was less than 40 m, which is consistent with the value obtained from the empirical formula. In the case of 20% CW, VHCF strength was not affected when the indent depth was less than 80 m. The results were much larger than the results obtained from the empirical formula and might have been caused by the plastic deformation, residual stress and probable deformation induced martensite transition around the indent.
Wan, T.; Naoe, Takashi; Kogawa, Hiroyuki; Futakawa, Masatoshi; Obayashi, Hironari; Sasa, Toshinobu
Materials, 12(4), p.681_1 - 681_15, 2019/02
Wakui, Takashi; Wakai, Eiichi; Naoe, Takashi; Kogawa, Hiroyuki; Haga, Katsuhiro; Takada, Hiroshi; Shintaku, Yohei*; Li, T.*; Kanomata, Kenichi*
Choompa Techno, 30(5), p.16 - 20, 2018/10
A mercury target vessel has been used for the spallation neutron source at J-PARC. It has a complicated multi-layered structure composed of a mercury target and a surrounding double-walled water shroud, which is assembled with thin plates (minimum thickness of 3 mm) by welding. Thus, welding inspection during the manufacturing process is important. We investigated the applicability of new ultrasonic inspections using specimens (thickness of 3 mm) with defects to improve the accuracy of welding inspection for the mercury target vessel. Immersion ultrasonic testing using a probe (frequency of 50 MHz) could detect a spherical defect with a diameter of 0.2 mm. The size was smaller than target value of 0.4 mm. The length of unwelded region estimated using the phased array ultrasonic testing corresponded with the actual length (0.8 - 1.5 mm).
Naoe, Takashi; Wakui, Takashi; Kogawa, Hiroyuki; Wakai, Eiichi; Haga, Katsuhiro; Takada, Hiroshi
Advanced Experimental Mechanics, 3, p.123 - 128, 2018/08
A mercury target vessel, composed of SUS316L, is used for the pulsed neutron source and is assembled via TIG welding. While in operation, the target vessel suffers ca. 10 loading cycles with a high strain rate of ca. 50 s because of the proton-beam-induced pressure waves in mercury. The gigacycle fatigue strength for solution annealed SUS316L stainless steels and its welded specimens were investigated through ultrasonic fatigue tests. The experimental results showed that an obvious fatigue limit was not observed at fewer than 10 cycles for the base metal. In the case of no weld defects observed via penetration tests, the fatigue strength of the removed-weld-bead specimen, in which the weld lines were arranged at the center of the specimen, appeared to be slightly higher than that of the base metal. By contrast, as-welded specimens with the weld bead intact showed apparent degradation of the fatigue strength owing to the stress concentration around the weld toe.
Naoe, Takashi; Xiong, Z.*; Futakawa, Masatoshi
Journal of Nuclear Materials, 506, p.12 - 18, 2018/08
A mercury target for neutron source (made of 316L SS) suffers not only proton and neutron radiation damage, but also cyclic impact stress caused by pressure waves. In the previous study, we carried out an ultrasonic fatigue test to investigate the gigacycle fatigue strength of 316L SS, concluding that specimen surface temperature rose abruptly more than 300C just before failure. In this study, to clarify the mechanism of the temperature rise, we measured temperature distribution with a thermography during the fatigue test. The experimental results showed that the temperature rose locally only at the crack tip and the peak position moved with the crack propagation. We also carried out a nonlinear structural analysis by LS-DYNA to estimate the temperature rise with strain energy of elements. The analytical result showed that the heat due to plastic deformation at the crack tip is dominant for the temperature rise rather than the friction between crack surface.
Naoe, Takashi; Wakui, Takashi; Kinoshita, Hidetaka; Kogawa, Hiroyuki; Haga, Katsuhiro; Harada, Masahide; Takada, Hiroshi; Futakawa, Masatoshi
Journal of Nuclear Materials, 506, p.35 - 42, 2018/08
A mercury target vessel made of 316L SS is damaged due to the cavitation caused by the pressure waves in mercury. Cavitation damage reduces the structural integrity of the target front, called "beam window", being major factor to determine the lifetime of target vessel. Aiming at mitigating the cavitation damage by faster mercury flow in narrow channel, we employed a target vessel with a double-walled structure at the beam window along with a gas microbubbles injection. After operating the double-walled target vessel with a beam power of 300 to 500 kW, we cut out the beam window using an annular cutter to examine the damage inside it, and found that damages with maximum pit depth of approximately 25 m distributed in a belt on the specimen facing narrow channel. Furthermore, numerical simulation result showed that the distribution of negative pressure period from beam injection to 1 ms was correlated with the damage distribution in the narrow channel. It was suggested that the cavitation induced by relatively short negative pressure period contributed to the damage formation.
Wakui, Takashi; Wakai, Eiichi; Naoe, Takashi; Shintaku, Yohei*; Li, T.*; Murakami, Kazuya*; Kanomata, Kenichi*; Kogawa, Hiroyuki; Haga, Katsuhiro; Takada, Hiroshi; et al.
Journal of Nuclear Materials, 506, p.3 - 11, 2018/08
The mercury target vessel is designed as multi-walled structure with thin wall (min. 3 mm), and assembled by welding. In order to estimate the structural integrity of the vessel, it is important to measure the defects in welding accurately. For nondestructive tests of the welding, radiographic testing is applicable but it is difficult to detect for some defect shapes. Therefore it is effective to do ultrasonic testing together with it. Because ultrasonic methods prescribed in JIS inspect on the plate with more than 6 mm in thickness, these methods couldn't be applied as the inspection on the vessel with thin walls. In order to develop effective method, we carried out measurements using some testing method on samples with small defect whose size is specified. In the case of the latest phased array method, measured value agreed with actual size. It was found that this method was applicable to detect defects in the thin-walled structure for which accurate inspection was difficult so far.
Haga, Katsuhiro; Kogawa, Hiroyuki; Wakui, Takashi; Naoe, Takashi; Takada, Hiroshi
Journal of Nuclear Science and Technology, 55(2), p.160 - 168, 2018/02
The mercury target vessel used for the spallation neutron source in J-PARC has multi-walled structure made of stainless steel type 316L, which comprises a mercury vessel and a water shroud. In 2015, water leak incidents from the water shroud occurred while the mercury target was operated with a proton beam power of 500 kW. Several investigations were conducted to identify the cause of failure. The results of the visual inspections, mockup tests, and analytical evaluations suggested that the water leak was caused by the combination of two factors. One was the diffusion bonding failure due to the large thermal stress induced by welding of the bolt head, which fixes the mercury vessel and the water shroud, during the fabrication process. The other was the thermal fatigue failure of the seal weld due to the repetitive beam trip during the operating period. These target failures point to the importance of eliminating initial defects from welding lines and to secure the rigidity and reliability of welded structures. The next mercury target was fabricated with an improved design which adopted parts of monolithic structure machined by wire EDM to reduce welding lines, and intensified inspections to eliminate the initial defects. The operation with the improved target is planned to be started in October 2017.
Takada, Hiroshi; Haga, Katsuhiro; Teshigawara, Makoto; Aso, Tomokazu; Meigo, Shinichiro; Kogawa, Hiroyuki; Naoe, Takashi; Wakui, Takashi; Oi, Motoki; Harada, Masahide; et al.
Quantum Beam Science (Internet), 1(2), p.8_1 - 8_26, 2017/09
At the Japan Proton Accelerator Research Complex (J-PARC), a pulsed spallation neutron source provides neutrons with high intensity and narrow pulse width to promote researches on a variety of science in the Materials and life science experimental facility. It was designed to be driven by the proton beam with an energy of 3 GeV, a power of 1 MW at a repetition rate of 25 Hz, that is world's highest power level. A mercury target and three types of liquid para-hydrogen moderators are core components of the spallation neutron source. It is still on the way towards the goal to accomplish the operation with a 1 MW proton beam. In this paper, distinctive features of the target-moderator-reflector system of the pulsed spallation neutron source are reviewed.
Kogawa, Hiroyuki; Naoe, Takashi; Futakawa, Masatoshi; Haga, Katsuhiro; Wakui, Takashi; Harada, Masahide; Takada, Hiroshi
Journal of Nuclear Science and Technology, 54(7), p.733 - 741, 2017/07
A mercury target system has been operated to produce neutron beams at the spallation neutron source in the Japan Proton Accelerator Research Complex (J-PARC). Pressure waves are generated in mercury by rapid heat generation at the time of high-intensity short-pulse proton beam injection. Not only they cause cyclic stress but also induce the cavitation damage on the target vessel made from type 316L stainless steel. Reduction of these pressure waves is very important issue to ensure enough lifetime of the target vessel. To solve the issue, we have been developing the technique to inject microbubbles into mercury. In this study, we installed a microbubble generator in the mercury target vessel, and investigate the effect of proton beam condition and the microbubbles on the pressure wave mitigation by measuring the displacement velocity of the target vessel with an in-situ diagnostic system. As a result, we observed that the peak displacement velocity of the target vessel decreased down to 1/3 and 2/3 for the injected gas fractions of 0.4% and 0.1%, respectively.