Nihon Hoshasen Anzen Kanri Gakkai-Shi, 16(2), P. 41, 2017/11
I summarized the outline of the oral presentation session where the author served as chairman.
Kurita, Yoshiyuki; Saegusa, Jun; Maeda, Satoshi
Nihon Hoshasen Anzen Kanri Gakkai-Shi, 15(2), p.180 - 185, 2016/11
In 2012, JAEA has established a laboratory in Fukushima-city to implement radioactivity analysis of environmental samples. Key nuclides measured in this laboratory are Cs and Cs (radiocesium). For the purpose, situation of radiocesium inflow into the laboratory building and its effects were investigated based on high-resolution gamma-spectrometry with germanium detectors. As a result, the surface density of radiocesium in the laboratory was found to be far below the minimum detectable activities of survey instruments routinely used for radiation control purposes.
Tokuyasu, Kayoko; Furuta, Sadaaki*; Kokubu, Yoko; Umeda, Koji
Nihon Hoshasen Anzen Kanri Gakkai-Shi, 15(1), p.80 - 87, 2016/07
An optically stimulated luminescence reader (Riso TL/OSL DA-20) was installed in Toki Research Institute of Isotope Geology and Geochronology (Toki-shi, Gifu Prefecture), Japan Atomic Energy Agency (JAEA) for dating the geological sample. An accumulated dose of the sample is obtained using the reader. Sealed beta source of strontium-90 is required to be mounted on the reader because repeated artificial irradiation is necessary for the accumulated dose estimation. However, there are not many introduction examples for the reader domestically, and the information as to radiation control of the reader is limited. We therefore report here the process of source loading on the reader and radiation control associated with the use of the source.
Miyauchi, Hideaki; Yoshitomi, Hiroshi; Sato, Yoshitaka; Takahashi, Fumiaki; Tachibana, Haruo; Kobayashi, Ikuo*; Suzuki, Akifumi*
Nihon Hoshasen Anzen Kanri Gakkai-Shi, 12(1), p.41 - 45, 2013/07
In the Japan Atomic Energy Agency (JAEA), exposures to fingertips can be significant in radiological decontamination at the facilities with mixture fields of beta and (X) rays. The radiation doses to fingertips have been measured by ring type dosemeters equipped with thermoluminescence dosemeters (TLD) in JAEA. We applied small Optically Stimulated Luminescence (OSL) elements to the ring type dosemeter, which has the advantages in the use for long term and repeating in dose measurements comparing to the TLDs. In this report, we introduce the outline and the dose evaluation method of the new ring type dosimeter which we applied.
Nagano, Tetsushi; Mitamura, Hisayoshi; Yanase, Nobuyuki; Naganawa, Hirochika; Yasuda, Kenichiro; Yoshida, Zenko; Yamaguchi, Hiroaki*
Nihon Hoshasen Anzen Kanri Gakkai-Shi, 11(2), p.139 - 145, 2012/11
An on-site monitoring method for radioactive cesium concentration in water using a cesium adsorption disk and a GM survey meter has been developed to rapidly and easily ascertain whether the water quality meets standards on radiological contaminants in water. In this method, both dissolved and suspended forms of radioactive cesium are collected on the cesium adsorption disk by means of filtration of a water sample. Beta counting rate of the disk is converted into radioactivity using a conservative calibration factor obtained here. The present on-site method was applied to monitoring of decontaminated water of an outdoor school swimming pool in Date city after Fukushima Dai-ichi Nuclear Power Plant accident.
Nihon Hoshasen Anzen Kanri Gakkai-Shi, 6(2), P. 111, 2007/11
no abstracts in English
Nihon Hoshasen Anzen Kanri Gakkai-Shi, 5(2), p.92 - 93, 2006/10
no abstracts in English