Ariyoshi, Gen; Obayashi, Hironari; Sasa, Toshinobu
Journal of Nuclear Science and Technology, 59(9), p.1071 - 1088, 2022/09
Electromagnetic induction method is one of the effective techniques for local velocity measurement in heavy liquid metals. Ricou and Vives' probe and Von Weissenfluh's probe are famous instrumentations using a permanent magnet. However, sensitivity and measurement volume of the probes show unexpected variation since demagnetization of the magnet is occurred by temperature increase up to the Curie temperature. In this study, electromagnetic probe incorporating a miniature electromagnet was newly developed to overcome such unexpected variation. The diameter and the length of the sensor was 6 mm and 155 mm, respectively. The sensitivity and the measurement volume of the probe were assessed by measurement of local velocity of flowing mercury in a square channel. To clarify the validity for the measured velocity profiles, numerical velocity profiles were calculated and compared with experiment. And the validity for the measured velocity profiles were confirmed by calculated result.
Naoe, Takashi; Kinoshita, Hidetaka; Wakui, Takashi; Kogawa, Hiroyuki; Haga, Katsuhiro
JAEA-Technology 2022-018, 43 Pages, 2022/08
In the liquid mercury target system for the pulsed spallation neutron source of Materials and Life science experimental Facility (MLF) at the Japan in the Japan Proton Accelerator Research Complex (J-PARC), cavitation that is generated by the high-energy proton beam-induced pressure waves, resulting severe erosion damage on the interior surface of the mercury target vessel. The erosion damage is increased with increasing the proton beam power, and has the possibility to cause the leakage of mercury by the penetrated damage and/or the fatigue failure originated from erosion pits during operation. To achieve the long term stable operation under high-power proton beam, the mitigation technologies for cavitation erosion consisting of surface modification on the vessel interior surface, helium gas microbubble injection, double-walled beam window structure has been applied. The damage on interior surface of the vessel is never observed during the beam operation. Therefore, after the target operation term ends, we have cut out specimen from the target nose of the target vessel to inspect damaged surface in detail for verification of the cavitation damage mitigation technologies and lifetime estimation. We have developed the techniques of specimen cutting out by remote handling under high-radiation environment. Cutting method was gradually updated based on experience in actual cutting for the used target vessel. In this report, techniques of specimen cutting out for the beam entrance portion of the target vessel in high-radiation environment and overview of the results of specimen cutting from actual target vessels are described.
Teshigawara, Makoto; Nakamura, Mitsutaka; Kinsho, Michikazu; Soyama, Kazuhiko
JAEA-Technology 2021-022, 208 Pages, 2022/02
The Materials and Life science experimental Facility (MLF) is an accelerator driven pulsed spallation neutron and muon source with a 1 MW proton beam. The construction began in 2004, and we started beam operation in 2008. Although problems such as exudation of cooling water from the target container have occurred, as of April 2021, the proton beam power has reached up to 700 kW gradually, and stable operation is being performed. In recent years, the operation experience of the rated 1 MW has been steadily accumulated. Several issues such as the durability of the target container have been revealed according to the increase in the operation time. Aiming at making a further improvement of MLF, we summarized the current status of achievements for the design values, such as accelerator technology (LINAC and RCS), neutron and muon source technology, beam transportation of these particles, detection technology, and neutron and muon instruments. Based on the analysis of the current status, we tried to extract improvement points for upgrade of MLF. Through these works, we will raise new proposals that promote the upgrade of MLF, attracting young people. We would like to lead to the further success of researchers and engineers who will lead the next generation.
Harada, Masahide; Teshigawara, Makoto; Oi, Motoki; Oikawa, Kenichi; Takada, Hiroshi; Ikeda, Yujiro
Nuclear Instruments and Methods in Physics Research A, 1000, p.165252_1 - 165252_8, 2021/06
This study explores high-energy neutron components of the extracted neutron beam at J-PARC pulsed neutron source using the foil activation method with threshold reactions. Foils of aluminum, gold, bismuth, niobium, and thulium were used to cover the neutron energy range from 0.3 MeV to 79.4 MeV. The experiment was performed using neutron beams of BL10 (NOBORU). The foils were irradiated by a neutron beam at 13.4 m from the moderator. To characterize high-energy neutron fields for irradiation applications, reaction rates in three different configurations with and without BC slit and Pb filter were examined. To compare the experiments with calculations given for the user, reaction rates for corresponding reactions were calculated by the PHITS code with the JENDL-3.2 and the JENDL dosimetry file. Although there was a systematic tendency in C/E (Calculation/Experiment) ratios for different threshold energies, which C/E ratio decreased as threshold energy increased up to 100 MeV, and all C/E ratios were in the range of 1.00.2. This indicated that high-energy neutron calculations were adequate for the analysis of experimental data for NOBORU users.
Kasugai, Yoshimi; Sato, Koichi; Takahashi, Kazutoshi*; Miyamoto, Yukihiro; Kai, Tetsuya; Harada, Masahide; Haga, Katsuhiro; Takada, Hiroshi
JPS Conference Proceedings (Internet), 33, p.011144_1 - 011144_6, 2021/03
A spallation neutron source with a mercury target has been in operation at the Materials and Life Science Experimental Facility of J-PARC since 2008. The target vessel made of stainless steel is required to be exchanged periodically due to radiation damage etc. In this presentation, tritium gas release observed in the first series of exchange work in 2011 and the analytical results will be shown.
Wakui, Takashi; Wakai, Eiichi; Kogawa, Hiroyuki; Naoe, Takashi; Hanano, Kohei*; Haga, Katsuhiro; Shimada, Tsubasa*; Kanomata, Kenichi*
Materials Science Forum, 1024, p.145 - 150, 2021/03
To realize a high beam power operation at the J-PARC, a mercury target vessel covered with water shroud was developed. In the first step, to realize an operation at 500 kW, the basic structure of the initial design was followed and the connection method between the mercury vessel and the water shroud was changed. Additionally, the operation at a beam power of 500 kW was realized in approximately eight months. In the second step, to realize the operation at 1 MW, the new structure in which only rear ends of vessels were connected was investigated. Cooling of the mercury vessel is used to reduce thermal stress and thick vessels of the water shroud are used to increase stiffness for the internal pressure; therefore, it was adopted. The stress in each vessel was lower than the allowable stress based on the pressure vessel code criteria prescribed in the Japan Industrial Standard, and confirmation was obtained that the operation with a beam power of 1 MW could be conducted.
Takada, Hiroshi; Haga, Katsuhiro
JPS Conference Proceedings (Internet), 28, p.081003_1 - 081003_7, 2020/02
At the Japan Proton Accelerator Research Complex (J-PARC), the pulsed spallation neutron source has been in operation with a redesigned mercury target vessel from October 2017 to July 2018, during which the operational beam power was restored to 500 kW and the operation with a 1-MW equivalent beam was demonstrated for one hour. The target vessel includes a gas-micro-bubbles injector and a 2-mm-wide narrow mercury flow channel at the front end as measures to suppress the cavitation damage. After the operating period, it was observed that the cavitation damage at the 3-mm-thick front end of the target vessel could be suppressed less than 17.5 m.
Katano, Ryota; Yamanaka, Masao*; Pyeon, C. H.*
Journal of Nuclear Science and Technology, 57(2), p.169 - 176, 2020/02
We proposed the linear combination method as a subcriticality measurement method which estimates the prompt neutron decay constant () correlated with the subcriticality using measurement results obtained at multiple detector positions. In the previous study, we confirmed applicability of the linear combination method through the pulsed neutron experiment with DT neutron source at Kyoto University Critical Assembly (KUCA). In this study, we conduct the pulsed neutron source experiment with spallation neutrons at KUCA and confirm the robustness of the linear combination to neutron sources.
Aso, Tomokazu; Tatsumoto, Hideki*; Otsu, Kiichi*; Kawakami, Yoshihiko*; Komori, Shinji*; Muto, Hideki*; Takada, Hiroshi
JAEA-Technology 2019-013, 77 Pages, 2019/09
At Materials and Life Science experimental Facility (MLF) of the Japan Proton Accelerator Research Complex (J-PARC), a 1-MW pulsed spallation neutron source is equipped with a cryogenic hydrogen system which circulates liquid hydrogen (20 K and 1.5 MPa) to convert high energy neutrons generated at a mercury target to cold neutrons at three moderators with removing nuclear heat of 3.8 kW deposited there. The cryogenic system includes an accumulator with a bellows structure in order to absorb pressure fluctuations generated by the nuclear heat deposition in the system. Welded inner bellows of the first accumulator was failured during operation, forcing us to improve the accumulator to have sufficient pressure resistance and longer life-time. We have developed elemental technologies for manufacturing welded bellows of the accumulator by a thick plate with high pressure resistance, succeeding to find optimum welding conditions. We fabricated a prototype bellows block and carried out an endurance test by adding a pressure change of 2 MPa repeatedly. As a result, the prototype bellows was successfully in use exceeding the design life of 10,000 times. Since distortions given during welding and assembling affect functionality and lifetime of the bellows, we set the levelness of each element of the bellows as within 0.1. The improved accumulator has already been in operation for about 25,000 hours as of January 2019, resulting that the number of strokes reached to 16,000. In July 2018, we demonstrated that the accumulator could suppress the pressure fluctuation generated by the 932 kW beam injection as designed. As current operational beam power is 500 kW, the current cryogenic hydrogen system could be applicable for stable operation at higher power in the future.
Wakui, Takashi; Ishii, Hideaki*; Naoe, Takashi; Kogawa, Hiroyuki; Haga, Katsuhiro; Wakai, Eiichi; Takada, Hiroshi; Futakawa, Masatoshi
Materials Transactions, 60(6), p.1026 - 1033, 2019/06
The mercury target has large size as 220.127.116.11 m. In view of reducing the amount of wastes, we studied the structure so that the fore part could be separated. The flange is required to have high seal performance less than 110 Pa m/s. Invar with low thermal expansion is a candidate. Due to its low stiffness, however, the flange may deform when it is fastened by bolts. Practically invar is reinforced with stainless steel where all interface between them has to be bonded completely with the HIP bonding. In this study, we made specimens at four temperatures and conducted tensile tests. The specimen bonded at 973 K had little diffusion layer, and so fractured at the interface. The tensile strength reduced with increasing the temperature, and the reduced amount was about 10% at 1473 K. The analyzed residual stresses near the interface increased by 50% at maximum. Then, we concluded that the optimum temperature was 1173 K.
Proceedings of 13th International Symposium on Advanced Science and Technology in Experimental Mechanics (13th ISEM'18) (USB Flash Drive), 6 Pages, 2018/10
Issues on the engineering technologies relating to high-power spallation neutron sources with liquid metals are introduced. The present status on research activities and results was reviewed.
Harada, Masahide; Teshigawara, Makoto; Oi, Motoki; Klinkby, E.*; Zanini, L.*; Batkov, K.*; Oikawa, Kenichi; Toh, Yosuke; Kimura, Atsushi; Ikeda, Yujiro
Nuclear Instruments and Methods in Physics Research A, 903, p.38 - 45, 2018/09
Wakui, Takashi; Wakai, Eiichi; Naoe, Takashi; Shintaku, Yohei*; Li, T.*; Murakami, Kazuya*; Kanomata, Kenichi*; Kogawa, Hiroyuki; Haga, Katsuhiro; Takada, Hiroshi; et al.
Journal of Nuclear Materials, 506, p.3 - 11, 2018/08
The mercury target vessel is designed as multi-walled structure with thin wall (min. 3 mm), and assembled by welding. In order to estimate the structural integrity of the vessel, it is important to measure the defects in welding accurately. For nondestructive tests of the welding, radiographic testing is applicable but it is difficult to detect for some defect shapes. Therefore it is effective to do ultrasonic testing together with it. Because ultrasonic methods prescribed in JIS inspect on the plate with more than 6 mm in thickness, these methods couldn't be applied as the inspection on the vessel with thin walls. In order to develop effective method, we carried out measurements using some testing method on samples with small defect whose size is specified. In the case of the latest phased array method, measured value agreed with actual size. It was found that this method was applicable to detect defects in the thin-walled structure for which accurate inspection was difficult so far.
Teshigawara, Makoto; Harada, Masahide; Oi, Motoki; Takada, Hiroshi
Journal of Physics; Conference Series, 1021(1), p.012061_1 - 012061_4, 2018/06
At the spallation neutron source of J-PARC, the structural material of moderators and reflector, such as an aluminum alloy, is going to reach to the design value (20 DPA) around 2020 by an accumulation of irradiation-damage. We started the fabrication of the spare moderators and reflector in 2013 with following design of two improvements. The invar joints, such as invar-A6061 and invar- SS316L joints were newly developed to utilize them in the cryogenic multi-layered pipe with 5th annular geometry, improving the fabrication procedure much simple. The Gold-Indium-Cadmium (Au-In-Cd) as a decoupler material is also developed to reduce residual radioactivity of the used components significantly for the decoupled moderator. In this presentation, we will report these results and progress of fabrication.
Plasma and Fusion Research (Internet), 13(Sp.1), p.2505013_1 - 2505013_8, 2018/03
The pulsed spallation neutron source of Japan Proton Accelerator Research Complex (J-PARC) has been supplying users with high intensity and sharp pulse cold neutrons using the moderators with following distinctive features; (1) 100% para-hydrogen for increasing pulse peak intensity with decreasing pulse tail, (2) cylindrical shape with 14 cm diam.12 cm long for providing high intensity neutrons to wide neutron extraction angles of 50.8, (3) neutron absorber made from Ag-In-Cd alloy to make pulse width narrower and pulse tails lower. Actually, it was measured at a low power operation that high neutron intensity of 4.510 n/cm/s/sr could be emitted from the coupled moderator surface for 1-MW operation, and a superior resolution of d/d = 0.035% was achieved at a beamline (BL8) with a poisoned moderator, where d is the d-spacing of reflection. Towards the goal to achieve the target operation at 1-MW for 5000 h in a year, technical developments to mitigate cavitation damages on the target vessel with injecting gas micro-bubbles into mercury target and design improvement of target vessel structure to reducing welds and bolt connections as much as possible are under way.
Teshigawara, Makoto; Ikeda, Yujiro; Oi, Motoki; Harada, Masahide; Takada, Hiroshi; Kakishiro, Masanori*; Noguchi, Gaku*; Shimada, Tsubasa*; Seita, Kyoichi*; Murashima, Daisuke*; et al.
Nuclear Materials and Energy (Internet), 14, p.14 - 21, 2018/01
We developed an Au-In-Cd (AuIC) decoupler material to reduce induced radioactivity instead of Ag-In-Cd one, which has a cut off energy of 1eV. In order to implement it into an actual moderator-reflector assembly, a number of critical engineering issues need to be resolved with regard to large-sized bonding between AuIC and A5083 alloys by the hot isostatic pressing process. We investigated this process in terms of the surface conditions, sizes, and heat capacities of large AuIC alloys. We also show a successful implementation of an AuIC decoupler into a reflector assembly, resulting in a remarkable reduction of radioactivity by AuIC compared to AIC without sacrificing neutronic performance.
Hamon, 27(4), p.155 - 158, 2017/11
The pulsed spallation neutron source driven by a high-power accelerator is one of the most powerful apparatus to provide high intensity and high quality neutrons with narrow pulse width for conducting cutting-edge researches in several domains of materials and life science. In this system, proton beams of several kW to MW order extracted from the high power accelerator is injected into a target, which is heavy metal, to generate vast amount of neutrons via the spallation reactions with the target nuclei, and slows down these neutrons to thermal to cold neutrons with a moderator and a reflector. Resultant neutron beams are then supplied to a suit of the state-of-the-art experimental devices. In this paper, mechanism to produce neutron beams and outline of the spallation neutron source, engineering design of a target system such as a mercury target, and technical topics to solve the pitting damage problem of the target vessel which is caused by the pressure wave of up to 40MPa at maximum generated in the mercury by the pulsed proton beam injection are reviewed by referring mainly to the mercury target system of the pulsed spallation neutron source at J-PARC.
Takada, Hiroshi; Haga, Katsuhiro; Teshigawara, Makoto; Aso, Tomokazu; Meigo, Shinichiro; Kogawa, Hiroyuki; Naoe, Takashi; Wakui, Takashi; Oi, Motoki; Harada, Masahide; et al.
Quantum Beam Science (Internet), 1(2), p.8_1 - 8_26, 2017/09
At the Japan Proton Accelerator Research Complex (J-PARC), a pulsed spallation neutron source provides neutrons with high intensity and narrow pulse width to promote researches on a variety of science in the Materials and life science experimental facility. It was designed to be driven by the proton beam with an energy of 3 GeV, a power of 1 MW at a repetition rate of 25 Hz, that is world's highest power level. A mercury target and three types of liquid para-hydrogen moderators are core components of the spallation neutron source. It is still on the way towards the goal to accomplish the operation with a 1 MW proton beam. In this paper, distinctive features of the target-moderator-reflector system of the pulsed spallation neutron source are reviewed.
Iwamoto, Hiroki; Nishihara, Kenji; Yagi, Takahiro*; Pyeon, C.-H.*
Journal of Nuclear Science and Technology, 54(4), p.432 - 443, 2017/04
Kasugai, Yoshimi; Harada, Masahide; Kai, Tetsuya; Oi, Motoki; Meigo, Shinichiro; Maekawa, Fujio
JAEA-Data/Code 2015-033, 28 Pages, 2016/03
The high-energy neutron fluxes and spectra around the mercury spallation neutron source at MLF of J-PARC were measured by the multi-foil activation method. The threshold energies of neutron reactions utilized in this experiment covered from 0.1 to 50 MeV. The foil irradiation was carried out on the first beam-run of MLF from May 30th to 31th, 2008. After the irradiation, the induced radioactivity of each foil was measured using an HPGe detector, and the neutron-induced reaction-rate distribution around the mercury target was determined. Using these data, the high-energy neutron fluxes and spectra were deduced with unfolding method in which the neutron spectra calculated with PHITS code were used as the initial-guess spectra. By comparison between the initial and the unfolded spectra, it was shown that most of the calculation results, which had been the basis of the neutronics design of the MLF target assembly, were consistent with the experimental data within 30%.