Kaburagi, Masaaki; Shimazoe, Kenji*; Terasaka, Yuta; Tomita, Hideki*; Yoshihashi, Sachiko*; Yamazaki, Atsushi*; Uritani, Akira*; Takahashi, Hiroyuki*
Nuclear Instruments and Methods in Physics Research A, 1046, p.167636_1 - 167636_8, 2023/01
We focus on the thickness and property controls of inorganic scintillators used for thermal neutron detection in intense -ray fields without considering pulse shape discrimination techniques. GS20 (a lithium glass) and LiCaAlF:Ce(LiCAF:Ce) cintillators with thicknesses of 0.5 and 1.0 mm, respectively, have been employed. Pulse signals generated by photomultiplier tubes, to which the scintillators were coupled, were inserted into a digital pulse processing unit with 1 Gsps, and the areas of waveforms were integrated for 360 ns. In a Co -ray field, the neutron detection for GS20 with a 0.5-mm thickness was possible at dose rates of up to 0.919 Gy/h; however, for LiCAF:Ce, neutron detection was possible at 0.473 Gy/h, and it failed at 0.709 Gy/h. Threfore, in a Co -ray field, the neutron/-ray discrimination of GS20 was better than that of LiCAF:Ce due to its better energy resolution and higher detection efficiency.
Endo, Shunsuke; Okudaira, Takuya*; Abe, Ryota*; Fujioka, Hiroyuki*; Hirota, Katsuya*; Kimura, Atsushi; Kitaguchi, Masaaki*; Oku, Takayuki; Sakai, Kenji; Shima, Tatsushi*; et al.
Physical Review C, 106(6), p.064601_1 - 064601_7, 2022/12
no abstracts in English
Yamamoto, Takeshi; Fujita, Manami; Gogami, Toshiyuki*; Harada, Takeshi*; Hayakawa, Shuhei*; Hosomi, Kenji; Ichikawa, Yudai; Ishikawa, Yuji*; Kamata, K.*; Kanauchi, H.*; et al.
EPJ Web of Conferences, 271, p.03001_1 - 03001_5, 2022/11
Kaburagi, Masaaki; Shimazoe, Kenji*; Kato, Masahiro*; Kurosawa, Tadahiro*; Takahashi, Hiroyuki*
Journal of Nuclear Science and Technology, 59(8), p.983 - 992, 2022/08
Kaburagi, Masaaki; Shimazoe, Kenji*; Kato, Masahiro*; Kurosawa, Tadahiro*; Kamada, Kei*; Kim, K. J.*; Yoshino, Masao*; Shoji, Yasuhiro*; Yoshikawa, Akira*; Takahashi, Hiroyuki*
Nuclear Instruments and Methods in Physics Research A, 1010, p.165544_1 - 165544_9, 2021/09
The number of nuclear facilities being decommissioned has been increasing worldwide, in particular following the accident of the Tokyo Electric Power Company Holdings' Fukushima Daiichi Nuclear Power Station in 2011. In these nuclear facilities, proper management of radioactive materials is required. Then, A -ray spectrometer with four segmentations using small volume CeBr scintillators with a dimension of was developed. The four scintillators were coupled to a multi-anode photomultiplier tube specific to intense radiation fields. We performed the -ray exposure study under Cs and Co radiation fields. Under the Cs radiation field, the relative energy resolution at 1375 mSv/h was the relative energy resolution at 1375 mSv/h was 9.20.05%, 8.00.08%, 8.00.03%, and 9.00.04% for the four channels, respectively.
Hirota, Takatoshi*; Nagoshi, Yasuto*; Hojo, Kiminobu*; Okada, Hiroshi*; Takahashi, Akiyuki*; Katsuyama, Jinya; Ueda, Takashi*; Ogawa, Takuya*; Yashirodai, Kenji*; Ohata, Mitsuru*; et al.
Proceedings of ASME 2021 Pressure Vessels and Piping Conference (PVP 2021) (Internet), 9 Pages, 2021/07
Fukuda, Tatsuya*; Takahashi, Ryo*; Hara, Takuhi*; Ohara, Koji*; Kato, Kazuo*; Matsumura, Daiju; Inaba, Yusuke*; Nakase, Masahiko*; Takeshita, Kenji*
Journal of Nuclear Science and Technology, 58(4), p.399 - 404, 2021/04
Sakakibara, Hiroshi; Aoki, Nobuhiro; Muto, Masahiro; Otabe, Jun; Takahashi, Kenji*; Fujita, Naoyuki*; Hiyama, Kazuhiko*; Suzuki, Hirokazu*; Kamogawa, Toshiyuki*; Yokosuka, Toru*; et al.
JAEA-Technology 2020-020, 73 Pages, 2021/03
The decommissioning is currently in progress at the prototype fast breeder reactor Monju. Fuel assemblies will be taken out of its core for the first step of the great task. Fuel assemblies stand on their own spike plugged into a socket on the core support plate and support with adjacent assemblies through their housing pads each other, resulting in steady core structure. For this reason, some substitutive assemblies are necessary for the purpose of discharging the fuel assemblies of the core. Monju side commissioned, therefore, Plutonium Fuel Development Center to manufacture the substitutive assemblies and the Center accepted it. This report gives descriptions of design, manufacture, and shipment in regard to the substitutive assemblies.
Nakano, Masanao; Fujii, Tomoko; Nemoto, Masashi; Tobita, Keiji; Seya, Natsumi; Nishimura, Shusaku; Hosomi, Kenji; Nagaoka, Mika; Yokoyama, Hiroya; Matsubara, Natsumi; et al.
JAEA-Review 2020-069, 163 Pages, 2021/02
Environmental radiation monitoring around the Tokai Reprocessing Plant has been performed by the Nuclear Fuel Cycle Engineering Laboratories, based on "Safety Regulations for the Reprocessing Plant of Japan Atomic Energy Agency, Chapter IV - Environmental Monitoring". This annual report presents the results of the environmental monitoring and the dose estimation to the hypothetical inhabitant due to the radioactivity discharged from the plant to the atmosphere and the sea during April 2019 to March 2020. In this report, some data include the influence of the accidental release from the Fukushima Daiichi Nuclear Power Station of Tokyo Electric Power Co., Inc. (the trade name was changed to Tokyo Electric Power Company Holdings, Inc. on April 1, 2016) in March 2011. Appendices present comprehensive information, such as monitoring programs, monitoring methods, monitoring results and their trends, meteorological data and discharged radioactive wastes. In addition, the data which were influenced by the accidental release and exceeded the normal range of fluctuation in the monitoring, were evaluated.
Kaburagi, Masaaki; Shimazoe, Kenji*; Kato, Masahiro*; Kurosawa, Tadahiro*; Kamada, Kei*; Kim, K. J.*; Yoshino, Masao*; Shoji, Yasuhiro*; Yoshikawa, Akira*; Takahashi, Hiroyuki*; et al.
Nuclear Instruments and Methods in Physics Research A, 988, p.164900_1 - 164900_8, 2021/02
An increasing number of nuclear facilities have been decommissioned worldwide following the 2011 accident of the TEPCO' Fukushima Daiichi Nuclear Power Station. During the decommissioning, radioactive materials have to be retrieved under proper management. In this study, a small cubic CeBr spectrometer with dimensions of 5 mm 5 mm 5 mm was manufactured to perform -ray spectroscopy under intense -ray fields. Furthermore, thanks to a fast digital process unit and a customized photomultiplier, the device could perform -ray spectroscopy at dose rates of over 1 Sv/h. The energy resolution (FWHM) at 662 keV ranged from 4.4% at 22 mSv/h to 5.2% at 1407 mSv/h for a Cs radiation field. Correspondingly, at 1333 keV, it ranged from 3.1% at 26 mSv/h to 4.2% at 2221 mSv/h for a Co radiation field, which suggested to realize -ray assessment of Cs, Cs, Co, and Eu at dose rates of over 1 Sv/h.
Okudaira, Takuya; Oku, Takayuki; Ino, Takashi*; Hayashida, Hirotoshi*; Kira, Hiroshi*; Sakai, Kenji; Hiroi, Kosuke; Takahashi, Shingo*; Aizawa, Kazuya; Endo, Hitoshi*; et al.
Nuclear Instruments and Methods in Physics Research A, 977, p.164301_1 - 164301_8, 2020/10
Kaburagi, Masaaki; Shimazoe, Kenji*; Otaka, Yutaka*; Uenomachi, Mizuki*; Kamada, Kei*; Kim, K. J.*; Yoshino, Masao*; Shoji, Yasuhiro*; Yoshikawa, Akira*; Takahashi, Hiroyuki*; et al.
Nuclear Instruments and Methods in Physics Research A, 971, p.164118_1 - 164118_8, 2020/08
Kawakita, Yukinobu; Kikuchi, Tatsuya*; Inamura, Yasuhiro; Tahara, Shuta*; Maruyama, Kenji*; Hanashima, Takayasu*; Nakamura, Mitsutaka; Kiyanagi, Ryoji; Yamauchi, Yasuhiro*; Chiba, Kaori*; et al.
Physica B; Condensed Matter, 551, p.291 - 296, 2018/12
There are elemental liquid metals with complex structures far from the hard sphere (HS) packing model. Liquid Bi has an asymmetric first peak in the structure factors S(Q). The pair distribution function g(r) exhibits strange distance ratio of 1:2 between the first and the second peaks. Since a HS model with two kinds of radius produces asymmetry of the main peak in S(Q), existence of short-lived covalent bonds was discussed. Contrarily, modulation of the atomic distribution by the Friedel oscillations of shielding electrons around metallic ions was discussed. To examine its bonding nature from viewpoints of dynamic correlation functions, we have measured neutron quasielastic scattering of liquid Bi by using cold disk chopper spectrometer installed at MLF of J-PARC. The van Hove function revealed that the shoulder structure located at a longer side of the first peak in g(r) exhibits a longer relaxation time than the main structures such as the first and second peaks.
Kaburagi, Masaaki; Sato, Yuki; Yoshihara, Yuri*; Shimazoe, Kenji*; Takahashi, Hiroyuki*; Torii, Tatsuo
Reactor Dosimetry; 16th International Symposium on Reactor Dosimetry (ISRD-16) (ASTM STP 1608), p.405 - 414, 2018/11
Kajimoto, Ryoichi; Nakamura, Mitsutaka; Inamura, Yasuhiro; Kamazawa, Kazuya*; Ikeuchi, Kazuhiko*; Iida, Kazuki*; Ishikado, Motoyuki*; Murai, Naoki; Kira, Hiroshi*; Nakatani, Takeshi; et al.
Journal of Physics; Conference Series, 1021(1), p.012030_1 - 012030_6, 2018/06
Pyeon, C. H.*; Vu, T. M.*; Yamanaka, Masao*; Sugawara, Takanori; Iwamoto, Hiroki; Nishihara, Kenji; Kim, S. H.*; Takahashi, Yoshiyuki*; Nakajima, Ken*; Tsujimoto, Kazufumi
Journal of Nuclear Science and Technology, 55(2), p.190 - 198, 2018/02
At the Kyoto University Critical Assembly, a series of reaction rate experiments is conducted on the accelerator-driven system (ADS) with spallation neutrons generated by the combined use of 100 MeV protons and a lead and bismuth target in the subcritical state. The reaction rates are measured by the foil activation method to obtain neutron spectrum information on ADS. Numerical calculations are performed with MCNP6.1 and JENDL/HE-2007 for high-energy protons and spallation process, JENDL-4.0 for transport and JENDL/D-99 for reaction rates. The reaction rates depend on subcriticality is revealed by the accuracy of the C/E (calculation/experiment) values. Nonetheless, the accuracy of the reaction rates at high-energy thresholds remains an important issue in the fixed-source calculations.
Onishi, Takashi; Sekioka, Ken*; Suto, Mitsuo*; Tanaka, Kosuke; Koyama, Shinichi; Inaba, Yusuke*; Takahashi, Hideharu*; Harigai, Miki*; Takeshita, Kenji*
Energy Procedia, 131, p.151 - 156, 2017/12
no abstracts in English
Kawamura, Seiko; Oku, Takayuki; Watanabe, Masao; Takahashi, Ryuta; Munakata, Koji*; Takata, Shinichi; Sakaguchi, Yoshifumi*; Ishikado, Motoyuki*; Ouchi, Keiichi*; Hattori, Takanori; et al.
Journal of Neutron Research, 19(1-2), p.15 - 22, 2017/11
Sample environment (SE) team at the Materials and Life Science Experimental Facility (MLF) in J-PARC has worked on development and operation of SE equipment and devices. All the members belong to one sub-team at least, such as Cryogenic and magnet, High temperature, High pressure, Soft matter and special environment including Pulse magnet, Hydrogen environment, Light irradiation and He spin filter. Cryostats, a magnet, furnaces, a VX-6-type Paris-Edinburgh press and a prototype of a Spin-Exchange Optical Pumping (SEOP) based He spin filter for polarized neutron beam experiments are in operation. Furthermore, a prototype of compact power supply for a pulsed magnet system is currently developed. In the J-PARC Research Building, several pieces of equipment for softmatter research such as a rheometer and a gas and vapor adsorption measurement instrument have been prepared.
Takahashi, Yoshio*; Fan, Q.*; Suga, Hiroki*; Tanaka, Kazuya; Sakaguchi, Aya*; Takeichi, Yasuo*; Ono, Kanta*; Mase, Kazuhiko*; Kato, Kenji*; Kanivets, V. V.*
Scientific Reports (Internet), 7(1), p.12407_1 - 12407_11, 2017/09
We focused on factors controlling partition of radiocesium on particulate matters and sediments in rivers of Fukushima and Chernobyl. Radiocesium is more soluble in the Pripyat River (Chernobyl) due to weaker interaction of radiocesium with clay minerals caused by the inhibition effect of the adsorbed humic substances. In contrast, particulate matters and sediments in the Kuchibuto River (Fukushima) display high adsorption affinity with lesser inhibition effect of adsorbed humic substances. This difference is possibly governed by the geology and soil type of provenances surrounding both catchments.
Tamura, Hajimu*; Shibata, Kenji*; Takahashi, Naoya; Niwa, Masakazu
JAEA-Testing 2017-001, 52 Pages, 2017/03
It is essential to understand the activity of faults in and around a target area to assess the long-term geosphere stability for geological isolation. Potassium-Argon (K-Ar) dating of fault gouge has been conducted in Tono Geoscience Center as one of the dating technique for faulting. This report includes methods for sample preparation, analyses, and age calculation on the K-Ar dating. This can contribute to an evaluation of calculated K-Ar ages.